'SOSORT consensus paper on brace action: TLSO biomechanics of correction (investigating the rationale for force vector selection)'
© Rigo et al; licensee BioMed Central Ltd. 2006
Received: 16 March 2006
Accepted: 20 July 2006
Published: 20 July 2006
The effectiveness of orthotic treatment continues to be controversial in international medical literature due to differences in the reported results and conclusions of various studies. Heterogeneity of the samples has been suggested as a reason for conflicting results. Besides the obvious theoretical differences between the brace concepts, the variability in the technical factors can also explain the contradictory results between same brace types. This paper will investigate the degree of variability among responses of scoliosis specialists from the Brace Study Ground of the International Society on Scoliosis Orthopedic and Rehabilitation Treatment SOSORT. Ultimately, this information could be a foundation for establishing a consensus and framework for future prospective controlled studies.
A preliminary questionnaire on the topic of 'brace action' relative to the theory of three-dimensional scoliosis correction and brace treatment was developed and circulated to specialists interested in the conservative treatment of adolescent idiopathic scoliosis. A particular case was presented (main thoracic curve with minor lumbar). Several key points emerged and were used to develop a second questionnaire which was discussed and full filed after the SOSORT consensus meeting (Milano, Italy, January 2005).
Twenty-one questionnaires were completed. The Chêneau brace was the most frequently recommended. The importance of the three point system mechanism was stressed. Options about proper pad placement on the thoracic convexity were divided 50% for the pad reaching or involving the apical vertebra and 50% for the pad acting caudal to the apical vertebra. There was agreement about the direction of the vector force, 85% selecting a 'dorso lateral to ventro medial' direction but about the shape of the pad to produce such a force. Principles related to three-dimensional correction achieved high consensus (80%–85%), but suggested methods of correction were quite diverse.
This study reveals that among participating SOSORT specialists there continues to be a strongly held and conflicting if not a contentious opinion regarding brace design and treatment. If the goal of a 'treatment consensus' is realistic and achievable, significantly more effort will be required to reconcile these differences.
Orthotic bracing is the most common non-surgical treatment for Adolescent Idiopathic Scoliosis (AIS), either alone or in combination with exercises. In spite of some negative reports [1–3], brace treatment has been shown to change the natural history of AIS  and reduce the incidence of surgery [5–7]. However, the mechanism of action by which braces prevent curve progression, is not well understood.
Generally speaking, bracing should unload the growth plates of the apical vertebral bodies on the concavity. Stokes has shown that an imposed vertebral deformity can be corrected by reversing the load used to create it . Therefore, the principles of the Hueter Volkmann law should be applicable to the correction of a scoliotic curve when there is sufficient residual growth. Nevertheless, evidence demonstrating that braces can affect the vertebral growth in spinal deformities is limited.
Design principles for many braces are contradictory and based on dated concepts, proposed and tested decades ago when the three-dimensional (3D) nature of AIS was rarely considered or incorporated into the brace design. The proper biomechanical principles for orthotic correction should apply derotational forces that correct in the coronal and axial plane in addition to producing normal spinal alignment in the sagittal plane .
In spite of numerous papers reporting on brace biomechanics [10–28], no principle, except perhaps 'the three point system', seems to be universally accepted. Many clinicians seem to fit braces empirically rather than using "curve-specific" biomechanical 3D models.
This paper will investigate the degree of variability among responses of scoliosis specialists from the Brace Study of the International Society on Scoliosis Orthopedic and Rehabilitation Treatment -SOSORT- (described below) regarding brace action and the selection of force vectors to effect optimal curve correction. Each participant was given the same patient scenario and curve pattern and was asked to complete a questionnaire indicating their treatment approach. Ultimately this information could be a foundation for establishing a consensus and framework for future prospective controlled studies.
Foundation for consensus
The formation of the SOSORT was a significant accomplishment of the Conference on Conservative management of Spinal Deformities, held January 2004 in Barcelona, Spain . At this meeting, participants agreed on the need for a prospective, multi-center, multi-national, controlled study using more specific bracing techniques that those reported with the Boston brace. They also agreed that a comprehensive patient outcome was a more important measure of success than the Cobb angle alone and this outcome should be based on assessment variables such as vertebral rotation, sagittal alignment, appearance of deformity, functional abilities and quality of life.
However, there was obvious disagreement, between different schools of thought, about how braces should correct a particular scoliosis pattern. Intensive discussion on brace action was postponed to future meetings, as participants sought a consensus before designing a prospective controlled study.
Following the formation of SOSORT in Barcelona, January 2004, a preliminary questionnaire was developed and circulated to all attendees and many specialists interested in the conservative treatment of adolescent idiopathic scoliosis. The simple questionnaire solicited their opinion on the topic of "brace action "where to push" and "why"" relative to the theory of 3-D scoliosis correction and brace treatment.
Twenty-one questionnaires were completed and returned for inclusion in this study. All but two of the responses recommended using a TLSO in treating the specified case. Of the two other responses, one did not answer this question and the other selected a Milwaukee brace. The most consistent selection was The Chêneau brace (13 times) and its variations. In addition, the Lyon brace was chosen three times and the modified Boston was chosen once.
The results of this study show the diversity of ideas and personal interpretations about the biomechanics of correction, brace design and treatment application in the proposed case. There was a wide range of treatment responses for this adolescent girl diagnosed with a classic idiopathic pattern of main right thoracic and minor left lumbar curvatures. Consensus was reached for three main biomechanical principles, at least from a theoretical point of view.
First, nearly everyone agreed on the importance of using the 'three point system' to correct the main thoracic curvature in the frontal plane. Principles related to 3D correction like 'derotation' and 'normalization of the sagittal profile', achieved greater consensus after the meeting (80%–85%) rather than before (50%). However, the suggested methods of correction were quite diverse.
When participants were asked which brace they preferred for this particular case, the Chêneau brace was the most frequently recommended. However, the variability in answers to specific questions on the biomechanics of correction indicates there are differing treatment methodologies. Consequently, the use of a particular name for a custom made brace, like the Chêneau, is no guaranty that there is a consistent standard in design or treatment.
A high thoracic extension on the concave thoracic side acts as the upper oppositional counter pad of the 'three point system' in this case. However, its effectiveness is enhanced or minimized by the pad placement and the direction of the force vector pushing the thoracic convexity. Over-correction of the shoulder level by using a left high thoracic counter pad was recommended by 50% of respondents, while the other 50% preferred a neutral, balanced alignment with no overcorrection. Several biomechanical studies have supported the use of the" three point system". Jonasson-Rajala et al  and later Perie et al  demonstrated that a high thoracic – axillary counter pad more effectively reduced both thoracic and lumbar curves rather than a lumbar pad combined with a low thoracic pad alone. Initial in brace correction, or primary correction, together with compliance, have been shown to be the most important factors predicting success of treatment . Thus, calculating the primary correction obtained could test the effectiveness of the 'three point system' applied for any particular brace. Further studies are necessary comparing correction and over-correction of the shoulder imbalance in order to recommend one or the other method.
Opinions about proper pad placement on the thoracic convexity were evenly split. Fifty percent opted for the pad reaching or involving the apical vertebra and the other 50% for the pad acting caudal to the apical vertebra but including the apical rib. However, 85% of respondents agreed upon the direction of the force vector, selecting the 'dorso lateral to ventro medial' direction. Both mechanisms, the placement of the pad and the direction of the force vector are important for the 'three points system' principle as well as for 3D correction principles.
The necessary combined forces to correct a thoracic scoliosis in 3D has been shown by Gignac D et al,  in a simulation of two new approaches comparing their results to the Boston brace. Different forces were applied at the thoracic apex level and the posterior displacement of the rib hump was blocked. An oblique force oriented 45°c degrees with respect to the frontal plane was added at the lumbar apex. The main force at thoracic level was applied with a dorsal direction on the ventral rib hump. A second main force pushed on the convexity with a latero medial direction. The suggested mechanism was able to reduce the Cobb angle in the frontal plane while maintaining the normal physiological curvatures in the sagittal plane, reducing axial rotation and rib hump. However, this new treatment approach must be personalized for each patient and still requires clinical evaluation.
In this consensus study the direction of the force vector pushing the thoracic convexity, which has been recommended for the majority of the participants, includes the lateral to medial direction. However it also includes a dorsal to ventral force, which would reduce the thoracic kyphosis. Designing a ventral pad pushing ventral to dorsal on the concave thoracic side could prevent this. This pad, oriented close to the frontal plane, in combination with a secondary dorsal pad on the convex side, oriented close to the sagittal plane, would be the suggested 3D mechanism. It effectively creates a 'pair of forces' to derotate in the axial plane and a coupling mechanism to increase the thoracic kyphosis.
Theoretically, this is the biomechanical principle of the Chêneau brace, the most recommended brace in this study. Despite this technical knowledge and preference for this specific brace, and though many of the specialists talk '3D correction', ironically, very few of them indicated an effective application of it. This could be the reason why the Chêneau brace has been reported to avoid thoracic flat back, for some authors [22, 31, 32], while for some others it increases the thoracic lordosis . This necessitates further study to clearly demonstrate that a particular brace can achieve an 'in vivo' 3D correction of a scoliotic curve.
Responses to the questions about abdominal pressure and the design of the pelvic section show not only a lack of agreement but contradictory design principles. Perhaps indicating that these are not essential elements in stabilizing the curvature in the proposed case or are concepts requiring further exploration and discussion before agreement is reached.
Although, all the SOSORT participants are recognized specialists in the treatment of scoliosis, having extensive knowledge of medical literature and significant clinical experience, many international opinions are missing from this study. Their inclusion would have increased the sample size and may have affected the outcome. Nevertheless, the objective of this study was not to establish new treatment guidelines as much as measure the level of consensus regarding current theoretical and treatment principles.
Through open discussion and completion of specific questionnaires a 'treatment consensus' was sought; a foundational consensus, essential for creating a future, prospective, multicenter, multinational study on bracing effectiveness. While the interactive process and discussions were constructive and some points of agreement were identified...was there sufficient progress towards an essential consensus?. Unfortunately, this study reveals that among participating SOSORT specialists there continues to be a strongly held and conflicting if not contentious opinion regarding TLSO design and treatment. If the goal of a 'treatment consensus' is realistic and achievable, significantly effort will be required to reconcile these differences.
The authors are very thankful to Luke Stikeleather, Virginia, for text editing.
- Noonan KJ, Weinstein SL, Jacobson WC, Dolan LA: Use of the Milwaukee brace for progressive idiopathic scoliosis. J Bone Joint Surg Am. 1996, 78: 557-567.PubMed
- Goldberg CJ, Fogarty EE, Hal JE, Emans JB: A statistical comparison between natural history of idiopathic scoliosis and brace treatment in skeletally immature adolescent girls. Spine. 1993, 18: 902-908.PubMedView Article
- Goldberg CJ, Moore DP, Fogarty EE: Adolescent idiopathic scoliosis: the effect of brace treatment on the incidence of surgery. Spine. 2001, 26: 42-47. 10.1097/00007632-200101010-00009.PubMedView Article
- Nachemson AL, Peterson LE, Members of the brace study group of the Scoliosis Research Society: Effectiveness of treatment with a brace in girls who have adolescent idiopathic scoliosis. A prospective, controlled study based on data from the brace study of the SRS. J Bone Joint Surg Am. 1995, 77: 815-822.PubMed
- Weiss HR, Weiss G, Schaar H: Conservative management in patients with scoliosis: does it reduce the incidence of surgery?. proceedings of the International Research Society of Spinal Deformities: 24–27 May 2002; Athens. Edited by: Grivas Th. 2002, 84-
- Maruyama T, Kitagawa T, Takeshita K, Mochizuki K, Nakamura K: Conservative treatment for adolescent idiopathic scoliosis: can it reduce the incidence of surgical treatment?. Pediatric Rahabilitation. 2003, 6: 215-219. 10.1080/13638490310001642748.
- Rigo M, Reiter CH, Weiss HR: Effect of conservative management on the prevalence of surgery in patients with adolescent idiopathic scoliosis. Pediatric Rehabilitation. 2003, 6: 209-214. 10.1080/13638490310001642054.PubMed
- Stokes IAF: Hueter-Volkmann effect. State of the Art Reviews. Spine. 2000, 14: 349-357.
- Dubousset J: Importance of the Three-dimensional concept in the treatment of scoliotic deformities. Proceedings of the Internationa Symposium on 3-D Scoliotic Deformities joined with the VII International Symposium on Spinal Deformity and Suface Topography. Edited by: Dansereau J. 1992, Montréal: Éditions del'École Polytechnique de Montréal, 302-311.
- Jonasson-Rajala E, Josefsson E, Lundberg B, Nilsson H: Boston thoracic brace in the treatment of idiopathic scoliosis. Initial correction. Clin Orthop. 1984, 183: 37-41.PubMed
- Labelle H, Dansereau J, Bellefleur C, Pitras B: 3D study of the immediate effect of the Boston brace on the scoliotic lumbar spine. Ann Chir. 1992, 46: 814-820.PubMed
- Labelle H, Dansereau J, Bellefleur C, Poitras B: Three-dimensional effect of the Boston brace on the thoracic spine and rib cage. Spine. 1996, 21: 59-64. 10.1097/00007632-199601010-00013.PubMedView Article
- Gignac D, Aubin CE, Dansereau J, Pulin F, labelle H: A biomechanical study of new orthotic treatment approaches for the 3D correction of scoliosis. Ann Chir. 1998, 52: 795-800.PubMed
- Wong MS, Evans JH: Biomechanical evaluation of the Milwaukee brace. Prosthet Orthop Int. 1998, 22: 54-67.
- Mouilleseaux B, Ebermeyer E, Charles C, Dauny G, Hecquet J, Graf H: 3D brace treatment: 3d immediate effect on thoracic, thoracolumbar and lumbar scoliotic curves. Technology and informatics 59. Research into Spinal Deformities 2. Edited by: Stokes IAF. 1999, Amsterdam IOS press, 354-357.
- Wong MS, Mak AF, Luk KD, Evans JH, Brown B: Effectiveness and biomechanics of spinal orthoses in the treatment of adolescent idiopathic scoliosis (AIS). Prosthet Orthop Int. 2000, 24: 148-162.View Article
- Gignac D, Aubin CE, Dansereau J, Labelle H: Optimization method for 3D bracing correction of scoliosis using a finite element model. Eur Spine J. 2000, 9: 185-190. 10.1007/s005860000135.PubMedPubMed CentralView Article
- Perie D, Sales De Gauzy J, Sevely A, Hobatho MC: In vivo geometrical evaluation of the Chêneau-Toulose Munster brace effect on scoliotic spine using MRI method. Clin Biomech. 2001, 16: 129-137. 10.1016/S0268-0033(00)00083-8.View Article
- Schmitz A, Kandyba J, Jaeger UE, Koenig R, Schmitt O: Brace effect in scoliosis in the sagittal plane-an MRI study. Z Orthop Ihre Grenzgeb. 2002, 140: 347-350. 10.1055/s-2002-32480.PubMedView Article
- Raso VJ, Lou E, Hill DL, Mahood JK, Moreau MJ: Is Boston brace mechanically effective in AIS?. Stud Health Technol Inform. 2002, 91: 378-382.PubMed
- Beausejour M, Petit Y, Grimard G, Aubin CE, Dansereau J, Labelle H: Relationship between strap tension, interface pressures and spine correction in brace treatment of scoliosis. Stud Health Technol Inform. 2002, 88: 207-211.PubMed
- Perie D, Sales De Gauzy J, Hobatho MC: Biomechanical evaluation of the Chêneau-Toulouse-Munster brace in the treatment of scoliosis using optimisation approach and finite element model. Med Biol Eng Comput. 2002, 40: 296-301. 10.1007/BF02344211.PubMedView Article
- Nicholson GP, Ferguson-Pell MW, Smith K, Edgar M, Morley T: The objective measurement of spinal orthosis for the treatment of idiopathic scoliosis. Spine. 2003, 28: 2243-2250. 10.1097/01.BRS.0000085098.69522.52.PubMedView Article
- Odermatt D, Mathieu PA, Beasejour M, Labelle H, Aubin CE: Electromyography of scoliosis patients treated with a brace. J Orthop Res. 2003, 21: 931-936. 10.1016/S0736-0266(03)00038-X.PubMedView Article
- Castro FP: Adolescent idiopathic scoliosis, bracing, and the Hueter-Volkmann principle. Spine J. 2003, 3: 180-185. 10.1016/S1529-9430(02)00557-0.PubMedView Article
- Perie D, Aubin CE, Petit Y, Beausejour M, Dansereau J, Labelle H: Boston brace correction in idiopathic scoliosis. Spine. 2003, 28: 1672-1677. 10.1097/00007632-200308010-00008.PubMed
- Perie D, Aubin CE, Lacroix M, Lafon Y, Labelle H: Biomechanical modelling of orthotic treatment of the scoliotic spine including a detailed representation of the brace-torso interface. Med Biol Eng Comput. 2004, 42: 339-344. 10.1007/BF02344709.PubMedView Article
- Schmitz A, Konig R, Kandyba J, Pennekamp P, Schmitt O, Jaeger UE: Visualization of the brace effect on the spinal profile in idiopathic scoliosis. Eur Spine J. 2005, 14: 138-143. 10.1007/s00586-004-0788-z.PubMedPubMed CentralView Article
- Rigo M: Abstracts of the international conference on conservative management of spinal deformities. Barcelona 23–24 January 2004. Pediatric Rehabilitation. 2004, 7: 51-64. 10.1080/13638490310001654736.
- Landauer F, Wimmer C, Behensky H: Estimating the final outcome of brace treatment for idiopathic thoracic scoliosis at 6-month follow-up. Pediatric Rehabilitation. 2003, 6: 201-207. 10.1080/13638490310001636817.PubMed
- Rigo M: 3D correction of trunk deformity in patients with idiopathic scoliosis using Chenau brace. Technology and Informatics 59. Research into spinal deformities 2. Edited by: Stokes IAF. 1999, Amsterdam:IOS Press, 362-365.
- Kotwicki T, Pietrzak S, Szulc A: Three-dimensional action of the Cheneau brace on thoracolumbar scoliosis. Research into Spinal Deformities 3. Edited by: Tanguy, peuchot. 2002, Amsterdam IOS Press, 226-229.
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.