Decompressive cervical laminectomy and lateral mass screw-rod arthrodesis. Surgical analysis and outcome
© Al Barbarawi et al; licensee BioMed Central Ltd. 2011
Received: 27 January 2011
Accepted: 19 May 2011
Published: 19 May 2011
This study evaluates the outcome and complications of decompressive cervical Laminectomy and lateral mass screw fixation in 110 cases treated for variable cervical spine pathologies that included; degenerative disease, trauma, neoplasms, metabolic-inflammatory disorders and congenital anomalies.
A retrospective review of total 785 lateral mass screws were placed in patients ages 16-68 years (40 females and 70 males). All cases were performed with a polyaxial screw-rod construct and screws were placed by using Anderson-Sekhon trajectory. Most patients had 12-14-mm length and 3.5 mm diameter screws placed for subaxial and 28-30 for C1 lateral mass. Screw location was assessed by post operative plain x-ray and computed tomography can (CT), besides that; the facet joint, nerve root foramen and foramen transversarium violation were also appraised.
No patients experienced neural or vascular injury as a result of screw position. Only one patient needed screw repositioning. Six patients experienced superficial wound infection. Fifteen patients had pain around the shoulder of C5 distribution that subsided over the time. No patients developed screw pullouts or symptomatic adjacent segment disease within the period of follow up.
decompressive cervical spine laminectomy and Lateral mass screw stabilization is a technique that can be used for a variety of cervical spine pathologies with safety and efficiency.
Keywordslateral mass arthrodesis cervical myelopathy spinal fixation decompressive laminectomy
Posterior cervical fixation with lateral mass screws was first introduced by Roy-Camille in 1979; it has been increasingly used since that time to treat a wide range of cervical spine disorders . posterior cervical fixation was frequently involved in form of wire and bone construct fixation. With a proven long-term effectiveness, and requires no special skills or x-ray guidance [2–4]. posterior cervical wire fixation may not be efficient in osteoporotic patient, as this technique can compromise the posterior cervical elements and may result in aggravating the primary pathology and worsen up the neurological status that requires full fixation by using the lateral mass fixation technique [5–7]. Furthermore, Stainless-steel wire can interfere with postoperative magnetic resonance (MR) imaging results, in contrast to the MRI compatible titanium screw/rod constructs. Lateral mass screw fixation has advantages over standard posterior wiring techniques; it can be done easily for many levels on patients with laminectomy and it can preserve the biomechanical forces. However, serious neural or vascular injury can explain the reservations of unfamiliar surgeons to this practice. so far, this method shows a global acceptance by many surgeons [8–10].
In this communication we reviewed and analyzed the surgical outcome of 110 consecutive cases treated with decompressive cervical Laminectomy and lateral mass fixation by using Anderson - Sekhon technique to deal with a punch of cervical disorders. Besides Operative and clinical outcomes; post operative computed tomography (CT) analysis is also provided with particular emphasis on clinical improvement and neurologic and vascular complications.
The Statistical Package for Social Sciences software (SPSS, version 15) was used for data processing and analysis. The subjects' variables were described using frequency distribution for categorical variables and mean and standard deviation for continuous variables. P value of ≤ 0.005 is considered not significant.
Clinical materials and methods
The study was approved by the ethical committee for human research (IRB) at Jordan University of Science and Technology. The study group consisted of 110 patients treated for multiple cervical pathologies performed in king Abdullah university hospital between Dec, 2005 and January, 2011. Decompressive cervical Laminectomy with a total of 785 lateral mass screws was applied in different cervical spine levels to deal with degenerative disease, spinal trauma (fracture-dislocation and hyperextension spinal cord injury ), cervical spine neoplasms, rheumatoid arthritis, calcification of the posterior longitudinal ligament and congenital anomalies. Exclusionary criteria included; Patients with soft tissue spinal cord tumours not affecting the bony elements or spinal instability, chronic or active infection. The severity of cervical myelopathy was assessed by using Nurick scale . The patient demographics were reviewed and analyzed in a retrospective manner.
The surgery was performed in a steady fashion. Fiberoptic Intubation was used as indicated. All cases were performed with digital fluoroscopic guidance. The lateral masses were initially drilled and tapped prior to laminectomy. Placement of screws was performed after cervical decompression. The entry point was about 1 mm medial to the midpoint of the lateral mass. The screws were angulated about 250 laterally and superiorly to achieve the best position of the lateral mass and to minimize the risk of neural or vascular violation, (modified Anderson and Sekhon techniques) [8, 12]. At C7 level, when its lateral mass is included in the fixation a more angulation was affected in comparison with other trajectories.
Intraoperatively, each screw position was assessed separately by imaging guidance before the final placement. For biological fusion; chips of auto-graft bone from the posterior elements or artificial bone were placed over the decorticated lateral masses and into the appropriate facet joints after screw insertion. Postoperatively all patient were placed into a hard nock collar and plain x-ray done on the first post operative day.
Postoperatively patients were evaluated clinically and radiologically at 4 weeks, 2 months, 6 months, 12 months, 24 months and 36 months. Follow-ups, in this study, ranged from 2 months to 3 years. All myelopathic patients discharged for rehabilitation programme.
shows the demographic distribution and the indications for decompressive Laminectomy and lateral mass fixation
Patient Demographics: (number = 110)
Cervical spine trauma
Central cord syndrome
Metabolic or inflammatory disorders
Intraoperatively, of the 785 lateral mass screws placed, there was no observation of vertebral artery injury or nerve root damage. Durotomy occurred in five cases that required intraoperative repair; all of those patients had a significant cervical stenosis. C7 was able to be adequately drilled with a steeper trajectory in seven cases. Poor screw placement occurred in about 14 screws from lateral mass breakout in patients with osteoporetic bone that required conversion to other trajectory. This was performed in 1.8% of all screws.
No active bleeding as a result of vertebral artery injury was noted in any case neither Post operatively, there was clinical evidence of vertebral artery injury as all patients were observed for local neck hematomas, vertebrobasilar stroke; and for any further neurological deterioration. However, some co morbidities were encountered; fifteen patients experienced a persistent C5 nerve root pain with a satisfactory postoperative CT scan showing no violation by screws of the C4-C5 neural foramen except in one female patient that required revision and her symptoms improved after revision.
shows the post operative surgical complications of the lateral mass fixation
Complications: (number = 110 cases, 785 screws)
Vertebral artery injury
Root Injury secondary to screws
Screw pullout or breakage(of 505 screws)
C5 root pain
Malposition that requires revision
Adjacent segment requiring surgery
Hematoma requiring evacuation
demonstrates the 405 lateral mass screw positions as evaluated by post op CT scan
Screw position assessed by CT scan
(number: 785 )
Violation of foramen transversarium by less than 1 mm.
Breaching of neural foramen
Entering the spinal canal
Violation facet joint
reveals the analysis of long term follow up
Long term follow up and Outcomes
Mean follow-up (mo)
Range of follow-up (mo)
Adjacent segment disease
Late vascular or neural damage
Related to instrumentation
In this study, age of patients with degenerative disc disease were approximated (P = .076) while it was variable for other pathologies. Patient's gender was male predominated (70%) (P = 0.311). The surgical principles for insertion of the lateral mass fixation was performed in the same manner. Additional surgical techniques were applied for vertebral body of the cervical spine tumours to achieve a gross total resection of the lesion and back up stabilization from anterior. In congenital anomalies of the cranio-cervical junction additional decompression of foramen magnum and incorporation of C2 pedicle was attained with satisfactory result. No neurovascular injury was observed. The vast majority of patients had a smooth operative and post operative courses. At the end of follow up for around 3 years the majority of survivors, basically the spondylotic, metabolic and congenital cases demonstrated a significant improvement. The co morbidities encountered in this study were not major and managed accordingly. Only one case developed a serious complication and she died of massive pulmonary embolism. Anther patient required adjustment of C5 screw because of persistent C5 radiculopathy and she improved after surgery.
This report documents that decompressive cervical laminectomy and lateral mass arthrodesis can be utilized for variable cervical spine pathologies with safety and efficiency. With emphasis on the use of Anderson- Sekhon trajectory as very reliable method and can provide adequate fixation with less violation of the adjacent structures.
Cervical spine is a place of many pathological lesions that could compromise its biomechanical stability. Restoration of stability may ultimately require fixation and placement of hard fixation devices. Posterior cervical spine stabilization is often administered to treat a variable cervical spine lesions that lead to spinal instability that include; cervical spondylotic degenerative disease, traumatic cervical spinal injury with and without fractures or with and without neurological deficits, metabolic -inflammatory lesions, primary and secondary neoplastic cervical spine lesions, infections, and in patients with previous wide cervical laminectomy without fixation. However, numerous surgical techniques and advances in spinal instrumentation have evolved over the last years. Lateral mass fixation has world widely gained popularity among spine surgeons with low morbidity and satisfactory outcome. Sekhon reported the largest series of subaxial lateral mass screw fixation with a total of 1024 screws and no related neuro-vascular injury observed [12, 13].
Many screw entry points and directions have been described since this technique was first introduced, Roy-Camille advocated the entry point of the screw is the midpoint of the lateral mass and the direction of the screw is to be perpendicular to the posterior aspect of the cervical spine and 100 outward . while Magerl proposed starting point is 2-3 mm medial and superior to the midpoint of the lateral mass and angling 300 superiorly and 250 laterally . Anderson recommended that the drilling point is 1 mm medial to the midpoint of the lateral mass and that the screw be angled 30-400 up and 100 lateral . An et al suggested angling 15-180. superiorly and 30-330 laterally, with a starting point 1 mm medial to the center of the lateral mass . Pait et al divided the lateral mass into four quadrants with the upper outer quadrant is the intention for screw insertion in this way its high likely to evade neurovascular injury . finally, sekhon recommended that by using Anderson's starting point and then angling 250 laterally and superiorly; this way is safe and easily applied. In regards to the lateral mass of C7 it can be attained with a steeper course without need for C7 pedicle .
Frequent clinical and cadaver investigations have been done on lateral mass fixation. Focusing on various trajectories to achieve proper placement of the screw and to avoid neural and vascular damage. Ebraheim et al on his cadaver study revealed the foramen transverarium is located in line with the midpoint of the lateral mass. So, the direction of the screw is to be laterally to avoid entry into the vertebral foramen [17, 18]. The work done by Xu et al reached that An technique is high likely to avoid neural damage compared to magerl and Anderson techniques [19, 20]. However, the incidence of nerve root violation when Roy-Camille or Magerl, Sekhon trajectories used is around 3.6%; this is most likely because of the lengthy screw and more lateral trajectory [13, 21, 22].
In terms screw length, Roy-Camille et al recommended 14-17 mm . An et al suggested a screw length of 11 mm is effective . Sekhon suggested that a 14-mm screw is safe and efficient based on the fact that the average vertical distance between the posterior midpoint of the lateral mass and the vertebral foramen from C3 to C6 is approximately 9-12 mm . As a result, insertion of a 14 mm screw obliquely should cross the lateral mass smoothly. In addition to that, a 14 mm screw can be bicorticate which adds further stability to the screw in place and causes no violation to the adjacent foramen, the Cadaveric studies of Heller et al concluded that bicorticate fixation with large diameter and non-self tapping screws had the utmost resistance to pullout [13, 23–25].
In comparison with other fixation techniques such as cervical pedicle screws, lateral mass fixation is safer, has higher success rate and low co-morbidities. In early studies, the failure rate was higher patients who underwent screw/plate constructs compared with the newer polyaxial screw/rod systems. The former systems were semi constricted with no cross link; which augment the stability of the system. In general, the newer polyaxial screw/rod systems are more constrained and essentially avoid screw pullout [25–27].
Wide Decompressive cervical Laminectomy with Lateral mass fixation by using Anderson - Sekhon trajectory is a safe and reliable surgical technique for posterior stabilization and proper for a wide range of cervical pathologies. With a long term follow up satisfactory results can be achieved. Neuro-vascular complication is usually low and is avoidable when using this trajectory.
- Roy-Camille R, Gaillant G, Bertreaux D: Early management of spinal injuries. Recent Advances Orthopedics. Edited by: McKibben B. 1979, Edinburgh: Churchill-Livingstone, 57-87.
- An HS, Coppes MA: Posterior cervical fixation for fracture and degenerative disc disease. Clin Orthop. 1997, X: 101-111.
- Crockard A: Evaluation of spinal laminar fixation by a new, flexible stainless steel cable (Sofwire): Early results. Neurosurgery. 1994, 35: 892-898. 10.1227/00006123-199411000-00013.PubMedView Article
- Geisler FH, Mirvis SE, Zrebeet H, Joslyn JN: Titanium wire internal fixation for stabilization of injury of the cervical spine: Clinical results and postoperative magnetic resonance imaging of the spinal cord. Neurosurgery. 1989, 25: 356-362. 10.1227/00006123-198909000-00006.PubMedView Article
- Deen HG, Birch BD, Wharen RE, Reimer R: Lateral mass screw-rod fixation of the cervical spine: A prospective clinical series with 1-year follow-up. Spine J. 2003, 3: 489-495.PubMed
- Horgan MA, Kellogg JX, Chesnut RM: Posterior cervical arthrodesis and stabilization: an early report using a novel lateral mass screw and rod technique. Neurosurgery. 1999, 44: 1267-1271.PubMed
- Muffoletto AJ, Hadjipavlou AG, Jensen RE, Nauta HJ, Necessary JT, Norcross-Nechay: Techniques and pitfalls of cervical lateral mass plate fixation. Am J Orthop. 2000, 29: 897-903.PubMed
- Anderson PA, Henley MB, Grady MS, Montesano P X, Winn H R: Posterior cervical arthrodesis with AO reconstruction plates and bone graft. Spine. 1991, 16: S72-S79. 10.1097/00007632-199103001-00012.PubMedView Article
- Shapiro S, Snyder W, Kaufman K, Abel T: Outcome of 51 cases of unilateral locked cervical facets: Interspinous braided cable for lateral mass plate fusion compared with interspinous wire and facet wiring with iliac crest. J Neurosurg. 1999, 91: 19-24.PubMed
- Ulrich C, Arand M, Nothwang J: Internal fixation on the lower cervical spine--Biomechanics and clinical practice of procedures and implants. Eur Spine J. 2001, 10: 88-100. 10.1007/s005860000233.PubMedPubMed CentralView Article
- Nurick S: The natural history and the results of surgical treatment of the spinal cord disorders associated with cervical spondylosis. Brain. 1972, 95: 101-108. 10.1093/brain/95.1.101.PubMedView Article
- Schultz KD, McLaughlin MR, Haid RW, Comey CH, Rodts GE, Alexander J: single staged anterior-posterior decompression and stabilization for complex cervical spine disorders. J Neurosurgery. 2000, 214-221. 2 supp
- Sekhon L: Posterior Cervical Lateral Mass Screw Fixation Analysis of 1026 Consecutive Screws in 143 Patients. J Spinal Disord Tech. 2005, 17:14:07
- Jeanneret B, Magerl F, Ward EH, Ward JC: Posterior stabilization of the cervical spine with hook plates. Spine. 1991, 16: S56-63. 10.1097/00007632-199103001-00010.PubMedView Article
- An HS, Gordin R, Renner K: Anatomic considerations for plate-screw fixation of the cervical spine. Spine. 1991, 16: S548-S551.PubMedView Article
- Pait TG, McAllister PV, Kaufman HH: Quadrant anatomy of the articular pillars (lateral cervical mass) of the cervical spine. J Neurosurg. 1995, 82: 1011-1014. 10.3171/jns.1918.104.22.1681.PubMedView Article
- Ebraheim NA, Xu R, Stanescu S, Yeasting RA: Anatomic relationship of the cervical nerves to the lateral masses. Am J Orthop. 1999, 28: 39-42.PubMed
- Ebraheim NA, Xu R, Yeasting RA: The location of the vertebral artery foramen and its relation to posterior lateral mass screw fixation. Spine. 1996, 21: 1291-1295. 10.1097/00007632-199606010-00002.PubMedView Article
- Xu R, Ebraheim NA, Klausner T, Yeasting RA: Modified Magerl technique of lateral mass screw placement in the lower cervical spine: An anatomic study. J Spinal Disord. 1998, 11: 237-240.PubMed
- Xu R, Haman SP, Ebraheim NA, Yeasting RA: The anatomic relation of lateral mass screws to the spinal nerves. A comparison of the Magerl, Anderson, and An techniques. Spine. 1999, 24: 2057-2061. 10.1097/00007632-199910010-00016.PubMedView Article
- Heller JG, Silcox DH, Sutterlin CE: Complications of posterior cervical plating. Spine. 1995, 20: 2442-2448. 10.1097/00007632-199511001-00013.PubMedView Article
- Wellman BJ, Follett KA, Traynelis VC: Complications of posterior articular mass plate fixation of the subaxial cervical spine in 43 consecutive patients. Spine. 1998, 23: 193-200. 10.1097/00007632-199801150-00009.PubMedView Article
- Muffoletto AJ, Yang j, Vadhva M, Hadjipavlou AG: Cervical instability with lateral mass plating: unicortical versus bicortical screw purchase. Spine. 2003, 28 (8): 778-81.PubMed
- Seybold EA, Baker JA, Criscitiello AA, Ordway NR, Park CK, Connolly PJ: Characteristics of unicortical and bicortical lateral mass screws in the cervical spine. Spine. 1999, 24: 2397-2403. 10.1097/00007632-199911150-00018.PubMedView Article
- Heller JG, Estes BT, Zaouali M, Diop A: Biomechanical study of screws in the lateral masses: Variables affecting pull-out resistance. J Bone Joint Surg Am. 1996, 78: 1315-1321.PubMed
- Abumi K, Kaneda K: Pedicle screw fixation for non-traumatic lesions of the cervical spine. Spine. 1997, 22: 1853-1863. 10.1097/00007632-199708150-00010.PubMedView Article
- Wu JC, Huang WC, Chen YC, Shih YH, Cheng H: Stabilization of subaxial cervical spines by lateral mass screw fixation with modified Magerl's technique. Surgical neurology. 2008, 25-33. Suppl 1:S1View Article
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.