Brace technology thematic series - The Sforzesco and Sibilla braces, and the SPoRT (Symmetric, Patient oriented, Rigid, Three-dimensional, active) concept
© Negrini et al; licensee BioMed Central Ltd. 2011
Received: 3 January 2011
Accepted: 9 May 2011
Published: 9 May 2011
Bracing is an effective strategy for scoliosis treatment, but there is no consensus on the best type of brace, nor on the way in which it should act on the spine to achieve good correction. The aim of this paper is to present the family of SPoRT (Symmetric, Patient-oriented, Rigid, Three-dimensional, active) braces: Sforzesco (the first introduced), Sibilla and Lapadula.
The Sforzesco brace was developed following specific principles of correction. Due to its overall symmetry, the brace provides space over pathological depressions and pushes over elevations. Correction is reached through construction of the envelope, pushes, escapes, stops, and drivers. The real novelty is the drivers, introduced for the first time with the Sforzesco brace; they allow to achieve the main action of the brace: a three-dimensional elongation pushing the spine in a down-up direction.
Brace prescription is made plane by plane: frontal (on the "slopes", another novelty of this concept, i.e. the laterally flexed sections of the spine), horizontal, and sagittal. The brace is built modelling the trunk shape obtained either by a plaster cast mould or by CAD-CAM construction. Brace checking is essential, since SPoRT braces are adjustable and customisable according to each individual curve pattern.
Treatment time and duration is individually tailored (18-23 hours per day until Risser 3, then gradual reduction). SEAS (Scientific Exercises Approach to Scoliosis) exercises are a key factor to achieve success.
The Sforzesco brace has shown to be more effective than the Lyon brace (matched case/control), equally effective as the Risser plaster cast (prospective cohort with retrospective controls), more effective than the Risser cast + Lyon brace in treating curves over 45 degrees Cobb (prospective cohort), and is able to improve aesthetic appearance (prospective cohort).
The SPoRT concept of bracing (three-dimensional elongation pushing in a down-up direction) is different from the other corrective systems: 3-point, traction, postural, and movement-based. The Sforzesco brace, being comparable to casting, may be the best brace for the worst cases.
Bracing is an effective strategy for scoliosis treatment, even if proof regarding its efficacy is currently still weak [1, 2]. Nevertheless, since the efficacy of bracing comes from both good quality construction and good compliance , bracing should never be interpreted only in terms of the brace applied, but also in terms of the management of patients . In fact, compliance is a characteristic neither of the treatment only, nor of the patient alone, but of the good interaction between these two factors and an expert treatment team able to reduce the burden of the brace and increase the coping abilities of the patient.
The expert members of the international Society on Scoliosis Orthopaedic and Rehabilitation Treatment (SOSORT) have not been able to reach a consensus on an optimal brace design, nor on the way it should act on the spine to achieve good correction ; on the contrary, they have reached consensus on the proper management of patients to achieve good results . Looking at the existing studies performed using the Scoliosis Research Society (SRS) methodological criteria, and dividing them into two groups (one respecting also the SOSORT criteria [6, 7], and another not doing so) it appears that the best results are obtained by the first group . So, the currently available international knowledge seems to agree that the type of brace used is less important than the way in which a brace is applied (SOSORT criteria) .
Nevertheless, this way of thinking could drive the field to a form of nihilism, where what you do (brace) is less important than how you do it (SOSORT criteria). Consequently, a comparison among the different tools applied by different physicians is mandatory, in order to understand these tools and to be able to separate their different indications. Until now, there have been very few comparison studies on different braces: one RCT , and some studies mainly with historical controls [10–16]. A critical assessment of some of these studies is vital, since in certain cases there has been doubt that the authors were experts in the use of the types of braces evaluated in the study. As a consequence, a more sound understanding of the basis behind the use of different braces is required to increase common background knowledge and to finally be able to safely compare the different instruments.
Three braces follow the SPoRT concept of correction. The Sforzesco brace (Figure 1) is constructed with rigid polycarbonate, in two pieces, connected posteriorly at the midline by a vertical aluminium bar and anteriorly by a closure that is rigid over the breast and below is made of soft inelastic bands. While the brace appears to be in full contact, in reality due to its symmetry and according to the theoretical body shape the patient would have without scoliosis, it provides space over depressions and pushes over pathological elevations.
Correction is reached through construction (shape of the envelope), pushes, drivers (concept newly-introduced with this brace), escapes, stops.
How to prescribe the brace and principles of correction
Prescribing the SPoRT braces requires a careful three-dimensional evaluation of the characteristics of the curve of each single patient. Clinical reasoning follows a systematic path by looking progressively at the single component of the deformity.
Frontal plane correction
Correction on the frontal plane is based on the identification not of the curves (as usual), but of the slopes, that are the most frontally flexed segments of the spine. In fact, since the brace works by pushing the spine from below, and due to the presence of the drivers that avoid undesired actions, pushes are focussed on the most severely flexed area of the spine (slopes). In a down-up direction, the following slopes can be described:
Low lumbar (Figure 10A): in a lumbar curve, below the apical vertebra.
High lumbar (Figure 10B): in a short thoracolumbar curve, below the apical vertebra; or in a very short lumbar curve, above the apical vertebra.
Lumbar (Figure 10C): in a wide thoracolumbar curve, below the apical vertebra.
Thoracolumbar (Figure 10D): in a lumbar curve, above the apical vertebra; or in a low thoracic curve, below the apical vertebra.
Thoracic (Figure 10E): in a thoracolumbar curve, above the apical vertebra; or in a single thoracic curve, below the apical vertebra; or in a double thoracic curve, above the apical vertebra of the distal curve and below the apical vertebra of the proximal one.
Distal thoracic (Figure 10F): specified only in Double Moe curves where three thoracic slopes are present, below the apical vertebra of the distal curve.
Proximal thoracic (Figure 10G): in a thoracic curve, above the apical vertebra.
Corrective action according to the frontal plane identified slope
shift of the trunk at the base
The whole trunk is shifted toward the concavity of the lumbar curve
elevation of the emithorax
The last ribs on the side of the convexity are elevated with a gradually decreasing compression in a down-up direction
shift of the trunk at the base and elevation of the emithorax
Combination of 1 and 2, on the same side
push on the distal ribs below the apical vertebra
On the side of convexity of the curve. All the ribs involved in the slope must be pushed. The rib corresponding to the apical vertebra is not involved (and sometimes also that below the apical)
push on the distal ribs above the apical vertebra
On the side of concavity of the curve
push on the distal ribs above the apical vertebra
On the side of concavity of the curve
elevation of the emithorax and push on the distal ribs below the apical vertebra
Combination of 2 and 4, on the same side
At the thoracic level, the ribs to be pushed must be identified, corresponding to the flexed vertebrae avoiding the apical vertebra.
The possible actions (not mutually exclusive) at the flanks include:
Shift: in the case of a low lumbar slope.
Stop: when there is a lumbar curve on the side opposite to the main slope.
Remodelling: to improve the aesthetics of one flattened flank.
On the frontal plane, the main drivers are placed laterally on the concave side, i.e. at the level of the waist and/or the thorax. They act mainly in a down-up direction from the apical vertebra of the curve, even if their action starts where the contra lateral push begins. They direct the forces above.
The horizontal plane
At the thoracic level, on the contrary, the derotatory push can damage the sagittal plane, and must be carefully planned. In this respect, it is mandatory not to reach the transverse processes, so as to allow for possible leverage by the ribs that could even result in a kyphosing action (Figure 15B). This leverage is at the base of the derotation and possibly deflexion action of the push on the hump. Moreover, the push must be below the apex of kyphosis to avoid its flattening. Above it, the push should be on the proximal counter-rotation appearing as a consequence of the thoracic thrust on the hump. This will allow on one hand an action to reconstruct the kyphosis, and on the other hand will increase the direct derotating (and modelling) push on the hump, as well as a realignment of the shoulder girdle otherwise rotated opposite to the convex side of the curve, due to the push on the thoracic hump.
At the thoracolumbar level, the action is usually similar to that at the lumbar one. In fact, most of the cases in this region appear with a junctional kyphosis, which is contrasted by a posterior push on the hump. In the few cases in which a junctional lordosis is present, the push must be present, but moderate to avoid increasing the sagittal deformity.
On the horizontal plane, the main drivers are anterior, where they avoid the anterior escape of the trunk driving in rotation, and posterior on the opposite side of the push, which are reached only when complete derotation is achieved and the push is driven upward.
The sagittal plane
This correction is almost completely done through the construction, since afterwards during checks it is almost impossible to really correct this point. The sagittal shaping of the brace during construction almost always changes according to the given patient's sagittal curve.
On the sagittal plane, all the drivers previously listed for the other planes play a crucial role in driving the forces not only upward but also slightly backward at the thoracolumbar junction, and anteriorly over the apex of kyphosis.
How to build the brace
The SPoRT concept always requires a customised construction of the brace according to the patient's individual requirements. CAD-CAM technologies usually allow us to obtain the best results, without using pre-built forms stored in databases, as is often done by others. Orthotists must directly shape the scanned trunk according to the patient's requirements. Once done, a final test must be made on the patient so as to change the first theoretical project and adapt it in the best possible way, depending on the real interaction between the body and the brace.
When the mould is ready, the plastic TLSO is fabricated, and the patient is fit according to his/her needs, allowing for good sitting position and total freedom of movement. The pushes are finally added at the level of the humps according to the desired corrections.
How to check the brace
Brace checking is a fundamental step in any brace construction . This is especially true in braces following the SPoRT concept, since they are adjustable and customisable according to any individual curve pattern. The reaction of the body to predisposed project of the brace should also be considered during prescription and building. Brace checking is moreover a key psychological intervention on the patient and family, mainly, but not only, with the first brace.
On the frontal plane, one has to search for the area in which correction is not ideal: corrections may be applied increasing the pushes or decreasing the drivers and counter pushes. On the sagittal plane, besides the appearance of the brace that must be properly aligned with respect to a normal kyphosis and lordosis, it is necessary to check inside the brace, and eventually either act on the posterior aluminium bar of the brace, or add plastazote pushes. On the horizontal plane, the check is made without the brace looking at the effect of the pushes on the humps. Finally, the total balance of the braced trunk is assessed, to avoid sagittal or frontal shifts (and rarely horizontal rotations). Other technical points to be checked include the non-overlap of pushes, that must be done on a plane by plane basis and the balance among the pushes (in the Sforzesco brace, pushing too much on a secondary curve has the consequence of reducing the efficacy on the main one).
An "in brace" radiograph is usually done only once, almost 45 days after the initial fitting of the first brace, but sometimes more often if there are problems.
Protocols and everyday usage
Brace treatment must almost always achieve very good aesthetic body shaping . It is intended to achieve radiographic results that are compatible with good functioning of the spine in adulthood, while the quality-of-life impact and psychological disturbances due to the brace must be minimised [5, 17, 37].
The type of brace is chosen according to the rigidity of the scoliosis to be treated. In large curves (over 40°), that are always rigid, the Sforzesco brace is used. Before puberty, in juveniles or infantile scoliosis patients, the Sibilla brace is prescribed with the very rare exceptions of a very rigid curve; in all other clinical situations, a case by case choice is made. The Lapadula brace is used as an alternative to the Sibilla in lumbar and thoraco-lumbar curves.
The goal of brace treatment varies according to the degree of curvature considered, and the forces that(in terms of rigidity of the brace and the hours of usage) are consequently administered . Treatment is tailored according to individual preferences, anthropometric characteristics and other risk factors such as rotation, hump, lumbar curve take-off, imbalance, etc. It usually starts at full time. Actually, the applied full-time concept varies between 18 and 23 hours per day [3, 38] with the goal of obtaining compliance. Treatment is carried out by wearing the brace at least 18 hours per day until the period of rapid growth is over and other adjustments due to the pathology are not foreseen. This is usually achieved at Risser stage 3.
We have recently shown in prospective controlled studies the importance of exercises in preparation for brace treatment so to increase its efficacy at first wearing , and in retrospective studies the usefulness of SEAS exercises in order to not lose correction while weaning from the brace . We have also shown which exercises are more useful in increasing the pushes of the brace .
Results and case reports
With specific studies we have shown that the Sforzesco brace:
is more effective than the Lyon brace after six months of treatment, with a matched case/control prospective study [10, 18] on 30 AIS patients aged 13 years and with curves of 38° Cobb on average: in the Sforzesco group 80% of patients improved and none worsened, while the Lyon group had results of 53% and 13%, respectively.
is equally effective as the Risser plaster cast to achieve the maximal correction after 18 months of treatment, with a prospective cohort study with a retrospective control group  on 41 AIS patients aged 14 years and with curves of 40° Cobb. The Sforzesco brace was shown to be more effective at reducing the thoracic curve, and its results were super imposable for the other regions. The Risser plaster brace was shown to be more effective on the thoracic hump and in regard to the cosmetic appearance of the flanks, but it also caused a serious reduction in kyphosis.
is more effective than the Risser cast + Lyon brace in treating curves over 45° Cobb at the end of growth, with a prospective cohort study in patients who utterly refused surgery  on 28 AIS patients aged 14 years and with curves of in a range of 45° to 58° Cobb. The patients braced with the Sforzesco had better results than those treated with the Risser cast in the thoracic curves, without any sagittal plane worsening. For the other parameters, the results were similar.
is able to improve aesthetics in scoliosis patients, with a prospective not-controlled cohort study  on 34 consecutive AIS patients 13 years old with curves of 32° Cobb with Aesthetic Index (AI)  scores of at least 5/6. At baseline, median AI was 6 (95% IC 5-6), but the score decreased to 3 (95% IC 0-5; p < 0.05) after six months with the brace, and this value was maintained in the 29 who completed the treatment (95% IC 1-6; p < 0.05 with respect to baseline).
The Sforzesco brace has been developed recently, but it is already one of the most tested TLSOs in the very weak scientific history of bracing. We are not able to compare it with any other that we did not use personally, but we can already state according to our results that its efficacy is higher than that of the Lyon brace , and comparable (or even higher as well) to that of the Risser cast . In fact, we use to think of the Sforzesco brace as a cast, with the great advantage on one hand that it can be removed to greatly increase patient comfort, and on the other hand that it can be used from the beginning to the end of treatment without problems, which cannot be done with the Risser cast. We cannot exclude in the future the possibility that the cast (or the Lyon brace) will still find a place in scoliosis treatment for some particular curves or patients, but we are not able now to exactly identify these clinical situations.
According to the reported results, we have a strong basis for reasoning that this brace could be more effective in the worst curves than other braces. In fact, to our knowledge, there is only one published paper with good results on curves over 45° Cobb, and they have been obtained either with Risser casting or with Sforzesco bracing . This conclusion needs to be supported by future evaluations and understanding, as well as study results reported by others with other braces.
Limitations can obviously be found today in the fact that the use of this brace is limited to Italy; we can anyway already state that the usage of the Sforzesco brace has already spread outside the first orthotic manufacturer and the first physician and his team. Nevertheless, we need studies from other teams, as is common with instruments at their first stages of development. A typical disadvantage of this instrument is that it is apparently simple. In fact, to a superficial observer it could appear as a simple full-contact brace. In reality, there are complex mechanical concepts and understanding that must be developed to be able to correctly apply this family of braces. Its apparent simplicity could easily drive its spread but could also lead to misconduct in its application. Moreover, another disadvantage is that the messages given to the patients are vital to success as well, and must be well understood. The SPoRT concept could also be applied to other braces beyond the ones presented here.
Looking at the braces used around the world, most of them are based on three-points systems, more or less three dimensional [26, 28, 53–61], but we can also recognise a traction system [62–64], a postural one [65–67], and finally a corrective-movement based [44, 68]. The SPoRT concept of bracing, due to its three-dimensional action of elongation pushing the spine in a down-up direction, is different from all the other corrective systems. The Sforzesco brace appears as the best brace for the worst curve magnitudes, being comparable to casting [16, 51, 69], with the obvious advantage of being removable and applicable for all duration of treatment.
Bracing is very hard work, in terms of conceptualisation of the practical work to be done, and of the interaction with the whole team, starting from the physician-orthotist relationship, to the physiotherapist, the patient and the family. It is a demanding, progressive, slow, artisanal effort in the art of patience. In this respect, it is quite the opposite of the short, one-shot, quick, highly demanding, current surgical fusion. As we use to say to our patients, bracing corresponds to the very slow pace of building oneself that humans usually have to face, contrary to the fast solution that he/she may tend to prefer and see as less demanding. Bracing in this respect also becomes a philosophy of one's approach to life, and this is one reason why it is difficult that the slow pace of a good conservative physician can also be the fast speed of a good orthopaedic surgeon, and vice versa. As well, there will always be patients who prefer bracing and others who prefer surgery. This relates to those with high degree curves; in low degree curves, the choice is between bracing and a "wait and see" strategy, applied in cases in which bracing is too demanding for that particular patient. But, in our own experience, at least in Italy, this is very rare , even if not avoidable.
We wish to thank Michele Romano for drawings; all our patients, who allowed us to increase our knowledge and skill; our collaborators who, working with us in team, were continuously part of what we learned and were able to do.
- Negrini S, Minozzi S, Bettany-Saltikov J, Zaina F, Chockalingam N, Grivas TB, Kotwicki T, Maruyama T, Romano M, Vasiliadis ES: Braces for idiopathic scoliosis in adolescents. Cochrane Database Syst Rev. 2010, CD006850-1Google Scholar
- Negrini S, Minozzi S, Bettany-Saltikov J, Zaina F, Chockalingam N, Grivas TB, Kotwicki T, Maruyama T, Romano M, Vasiliadis ES: Braces for idiopathic scoliosis in adolescents. Spine (Phila Pa 1976). 2010, 35 (13): 1285-1293.View ArticleGoogle Scholar
- Landauer F, Wimmer C, Behensky H: Estimating the final outcome of brace treatment for idiopathic thoracic scoliosis at 6-month follow-up. Pediatr Rehabil. 2003, 6 (3-4): 201-207.View ArticlePubMedGoogle Scholar
- Negrini S, Grivas TB, Kotwicki T, Rigo M, Zaina F: Guidelines on "Standards of management of idiopathic scoliosis with corrective braces in everyday clinics and in clinical research": SOSORT Consensus 2008. Scoliosis. 2009, 4 (1): 2-View ArticlePubMedPubMed CentralGoogle Scholar
- Rigo M, Negrini S, Weiss H, Grivas T, Maruyama T, Kotwicki T: 'SOSORT consensus paper on brace action: TLSO biomechanics of correction (investigating the rationale for force vector selection)'. Scoliosis. 2006, 1: 11-10.1186/1748-7161-1-11.View ArticlePubMedPubMed CentralGoogle Scholar
- Negrini S, Atanasio S, Fusco C, Zaina F: Effectiveness of complete conservative treatment for adolescent idiopathic scoliosis (bracing and exercises) based on SOSORT management criteria: results according to the SRS criteria for bracing studies - SOSORT Award 2009 Winner. Scoliosis. 2009, 4: 19-10.1186/1748-7161-4-19.View ArticlePubMedPubMed CentralGoogle Scholar
- Aulisa AG, Guzzanti V, Galli M, Perisano C, Falciglia F, Aulisa L: Treatment of thoraco-lumbar curves in adolescent females affected by idiopathic scoliosis with a progressive action short brace (PASB): assessment of results according to the SRS committee on bracing and nonoperative management standardization criteria. Scoliosis. 2009, 4: 21-10.1186/1748-7161-4-21.View ArticlePubMedPubMed CentralGoogle Scholar
- Janicki JA, Poe-Kochert C, Armstrong DG, Thompson GH: A comparison of the thoracolumbosacral orthoses and providence orthosis in the treatment of adolescent idiopathic scoliosis: results using the new SRS inclusion and assessment criteria for bracing studies. J Pediatr Orthop. 2007, 27 (4): 369-374. 10.1097/01.bpb.0000271331.71857.9a.View ArticlePubMedGoogle Scholar
- Wong MS, Cheng JC, Lam TP, Ng BK, Sin SW, Lee-Shum SL, Chow DH, Tam SY: The effect of rigid versus flexible spinal orthosis on the clinical efficacy and acceptance of the patients with adolescent idiopathic scoliosis. Spine. 2008, 33 (12): 1360-1365. 10.1097/BRS.0b013e31817329d9.View ArticlePubMedGoogle Scholar
- Negrini S, Marchini G: Efficacy of the symmetric, patient-oriented, rigid, three-dimensional, active (SPoRT) concept of bracing for scoliosis: a prospective study of the Sforzesco versus Lyon brace. Eura Medicophys. 2007, 43 (2): 171-181. discussion 183-174PubMedGoogle Scholar
- Weiss HR, Weiss GM: Brace treatment during pubertal growth spurt in girls with idiopathic scoliosis (IS): a prospective trial comparing two different concepts. Pediatr Rehabil. 2005, 8 (3): 199-206.View ArticlePubMedGoogle Scholar
- Katz DE, Richards BS, Browne RH, Herring JA: A comparison between the Boston brace and the Charleston bending brace in adolescent idiopathic scoliosis. Spine. 1997, 22 (12): 1302-1312. 10.1097/00007632-199706150-00005.View ArticlePubMedGoogle Scholar
- Howard A, Wright JG, Hedden D: A comparative study of TLSO, Charleston, and Milwaukee braces for idiopathic scoliosis. Spine. 1998, 23 (22): 2404-2411. 10.1097/00007632-199811150-00009.View ArticlePubMedGoogle Scholar
- Bunnell WP, MacEwen GD, Jayakumar S: The use of plastic jackets in the non-operative treatment of idiopathic scoliosis. Preliminary report. J Bone Joint Surg Am. 1980, 62 (1): 31-38.PubMedGoogle Scholar
- Climent JM, Sanchez J: Impact of the type of brace on the quality of life of Adolescents with Spine Deformities. Spine. 1999, 24 (18): 1903-1908. 10.1097/00007632-199909150-00007.View ArticlePubMedGoogle Scholar
- Negrini S, Atanasio S, Negrini F, Zaina F, Marchini G: The Sforzesco brace can replace cast in the correction of adolescent idiopathic scoliosis: A controlled prospective cohort study. Scoliosis. 2008, 3 (1): 15-10.1186/1748-7161-3-15.View ArticlePubMedPubMed CentralGoogle Scholar
- Negrini S, Marchini G, Tomaello L: Efficacy of the Symmetric, Patient-oriented, Rigid, Three-Dimensional (SPoRT) concept of bracing for scoliosis: a pair-controlled retrospettive short-term study on the Sforzesco Brace. 3rd International SOSORT Meeting: 2006 April 07-08 2006; Poznan, Poland: SOSORT. 2006Google Scholar
- Negrini S, Marchini G, Tomaello L: The Sforzesco brace and SPoRT concept (Symmetric, Patient-oriented, Rigid, Three-dimensional) versus the Lyon brace and 3-point systems for bracing idiopathic scoliosis. Stud Health Technol Inform. 2006, 123: 245-249.PubMedGoogle Scholar
- Sibilla P: Trent'anni di scoliosi. Lezione "non" magistrale. Rachide & Riabilitazione 2002. Edited by: Negrini S, Rainero G. 2002, Vigevano: Gruppo di Studio Scoliosi e patologie vertebrali, 1: 73-92.Google Scholar
- Sibilla P: Il trattamento conservativo attivo della scoliosi idiopatica in Italia. Le deformità vertebrali: stato dell'arte. Edited by: Negrini S, Sibilla P. 2001, Vigevano: Gruppo di Studio Scoliosi e patologie vertebrali, 2: 20-41.Google Scholar
- Risser JC: Scoliosis treated by cast correction and spine fusion. Clin Orthop Relat Res. 1976, 86-94. 116Google Scholar
- Mammano S, Scapinelli R: Plaster casts for the correction of idiopathic scoliosis. Acta Orthop Belg. 1992, 58 (Suppl 1): 81-84.PubMedGoogle Scholar
- Stagnara P: Les deformations du rachis. 1976, Paris: Expansion Scientifique FrancaiseGoogle Scholar
- Blount WP, Schmidt A: The Milwaukee brace in the treatment of scoliosis. J Bone Joint Surg. 1957, 37: 693-Google Scholar
- Moe JH: Indications for Milwaukee brace non-operative treatment in idiopathic scoliosis. Clin Orthop Relat Res. 1973, 38-43. 93Google Scholar
- Weiss HR, Rigo M: The cheneau concept of bracing--actual standards. Stud Health Technol Inform. 2008, 135: 291-302.PubMedGoogle Scholar
- Kotwicki T, Cheneau J: Passive and active mechanisms of correction of thoracic idiopathic scoliosis with a rigid brace. Stud Health Technol Inform. 2008, 135: 320-326.PubMedGoogle Scholar
- Rigo M, Weiss HR: The Cheneau concept of bracing--biomechanical aspects. Stud Health Technol Inform. 2008, 135: 303-319.PubMedGoogle Scholar
- Kotwicki T, Cheneau J: Biomechanical action of a corrective brace on thoracic idiopathic scoliosis: Cheneau 2000 orthosis. Disabil Rehabil Assist Technol. 2008, 3 (3): 146-153. 10.1080/17483100801905744.View ArticlePubMedGoogle Scholar
- Rigo MD, Villagrasa M, Gallo D: A specific scoliosis classification correlating with brace treatment: description and reliability. Scoliosis. 2009, 5 (1): 1-View ArticleGoogle Scholar
- Negrini S: The Evidence-Based ISICO Approach to Spinal Deformities. 2007, Milan, Boston: ISICO, 1Google Scholar
- Negrini S: Bracing adolescent idiopathic scoliosis today. Disabil Rehabil Assist Technol. 2008, 3 (3): 107-111. 10.1080/17483100801942887.View ArticlePubMedGoogle Scholar
- Negrini S, Grivas TB, Kotwicki T, Rigo M, Zaina F: Guidelines on "Standard of management of idiopathic scoliosis with corrective braces in everyday clinics and in clinical research": SOSORT Consensus 2008. Scoliosis. 2009, 4 (1): 2-View ArticlePubMedPubMed CentralGoogle Scholar
- Negrini S, Zaina F, Negrini F, Marchini G, Aulisa AG: Sforzesco brace (SPoRT Concept) versus Risser cast in adolescent idiopathic scoliosis treatment: similar efficacy, with reduced spinal side effects for the brace. 4th International Conference on Conservative Management of Spinal Deformities: 13-16 May 2007 2007. 2007, Boston: SOSORT (Society on Scoliosis Orthopaedic and Rehabilitation Treatment)Google Scholar
- Bunge EM, de Bekker-Grob EW, van Biezen FC, Essink-Bot ML, de Koning HJ: Patients' preferences for scoliosis brace treatment: a discrete choice experiment. Spine (Phila Pa 1976). 2010, 35 (1): 57-63. 10.1097/BRS.0b013e3181bdeaa6.View ArticleGoogle Scholar
- Zaina F, Negrini S, Fusco C, Atanasio S: How to improve aesthetics in patients with Adolescent Idiopathic Scoliosis (AIS): a SPoRT brace treatment according to SOSORT management criteria. Scoliosis. 2009, 4: 18-10.1186/1748-7161-4-18.View ArticlePubMedPubMed CentralGoogle Scholar
- Negrini S, Grivas TB, Kotwicki T, Maruyama T, Rigo M, Weiss HR: Why do we treat adolescent idiopathic scoliosis? What we want to obtain and to avoid for our patients. SOSORT 2005 Consensus paper. Scoliosis. 2006, 1: 4-10.1186/1748-7161-1-4.View ArticlePubMedPubMed CentralGoogle Scholar
- Wiley JW, Thomson JD, Mitchell TM, Smith BG, Banta JV: Effectiveness of the boston brace in treatment of large curves in adolescent idiopathic scoliosis. Spine. 2000, 25 (18): 2326-2332. 10.1097/00007632-200009150-00010.View ArticlePubMedGoogle Scholar
- Duval-Beaupere G, Lespargot A, Grossiord A: Flexibility of scoliosis. What does it mean? Is this terminology appropriate?. Spine. 1985, 10 (5): 428-432. 10.1097/00007632-198506000-00005.View ArticlePubMedGoogle Scholar
- Torell G, Nachemson A, Haderspeck-Grib K, Schultz A: Standing and supine Cobb measures in girls with idiopathic scoliosis. Spine. 1985, 10 (5): 425-427. 10.1097/00007632-198506000-00004.View ArticlePubMedGoogle Scholar
- Smania N, Picelli A, Romano M, Negrini S: Neurophysiological basis of rehabilitation of adolescent idiopathic scoliosis. Disabil Rehabil. 2008, 30 (10): 763-771. 10.1080/17483100801921311.View ArticlePubMedGoogle Scholar
- Stokes IA, Burwell RG, Dangerfield PH: Biomechanical spinal growth modulation and progressive adolescent scoliosis - a test of the 'vicious cycle' pathogenetic hypothesis: Summary of an electronic focus group debate of the IBSE. Scoliosis. 2006, 1: 16-10.1186/1748-7161-1-16.View ArticlePubMedPubMed CentralGoogle Scholar
- Burwell RG, Aujla RK, Grevitt MP, Dangerfield PH, Moulton A, Randell TL, Anderson SI: Pathogenesis of adolescent idiopathic scoliosis in girls - a double neuro-osseous theory involving disharmony between two nervous systems, somatic and autonomic expressed in the spine and trunk: possible dependency on sympathetic nervous system and hormones with implications for medical therapy. Scoliosis. 2009, 4: 24-10.1186/1748-7161-4-24.View ArticlePubMedPubMed CentralGoogle Scholar
- Coillard C, Leroux MA, Zabjek KF, Rivard CH: SpineCor--a non-rigid brace for the treatment of idiopathic scoliosis: post-treatment results. Eur Spine J. 2003, 12 (2): 141-148.PubMedGoogle Scholar
- Didier J: La plasticité de la fonction motrice. 2004, Paris: Springer, 1Google Scholar
- Zaina F, Negrini S, Atanasio S, Fusco C, Romano M, Negrini A: Specific exercises performed in the period of brace weaning can avoid loss of correction in Adolescent Idiopathic Scoliosis (AIS) patients: Winner of SOSORT's 2008 Award for Best Clinical Paper. Scoliosis. 2009, 4 (1): 8-10.1186/1748-7161-4-8.View ArticlePubMedPubMed CentralGoogle Scholar
- Nachemson AL, Peterson LE: Effectiveness of treatment with a brace in girls who have adolescent idiopathic scoliosis. A prospective, controlled study based on data from the Brace Study of the Scoliosis Research Society. J Bone Joint Surg Am. 1995, 77 (6): 815-822.PubMedGoogle Scholar
- Stagnara P, Mollon G, De Mauroy J: Reeducation des scolioses. 1990, Paris: Expansion Scientifique FrancaiseGoogle Scholar
- Romano M, Carabalona R, Petrilli S, Sibilla P, Negrini S: Forces exerted during exercises by patients with adolescent idiopathic scoliosis wearing fiberglass braces. Scoliosis. 2006, 1: 12-10.1186/1748-7161-1-12.View ArticlePubMedPubMed CentralGoogle Scholar
- Negrini S, Negrini A, Romano M, Verzini N, Parzini S: A controlled prospective study on the efficacy of SEAS.02 exercises in preparation to bracing for idiopathic scoliosis. Stud Health Technol Inform. 2006, 123: 519-522.PubMedGoogle Scholar
- Negrini S, Negrini F, Fusco C, Zaina F: Idiopathic scoliosis patients with curves over 45 Cobb degrees refusing surgery can be effectively treated through bracing with curve improvements. The Spine Journal. 2011,Google Scholar
- Zaina F, Negrini S, Atanasio S: TRACE (Trunk Aesthetic Clinical Evaluation), a routine clinical tool to evaluate aesthetics in scoliosis patients: development from the Aesthetic Index (AI) and repeatability. Scoliosis. 2009, 4 (1): 3-10.1186/1748-7161-4-3.View ArticlePubMedPubMed CentralGoogle Scholar
- Cheneau J: Corset-Cheneau. Manuel d'Orthopèdie des scolioses suivant la technique originale; 1. 1994, Paris: Frison RocheGoogle Scholar
- Weiss HR, Werkmann M, Stephan C: The ScoliOlogiC "Cheneau light" brace--does the reduction of material affect the desired correction?. Stud Health Technol Inform. 2006, 123: 250-254.PubMedGoogle Scholar
- Bassett GS, Bunnell WP, MacEwen GD: Treatment of idiopathic scoliosis with the Wilmington brace. Results in patients with a twenty to thirty-nine-degree curve. J Bone Joint Surg Am. 1986, 68 (4): 602-605.PubMedGoogle Scholar
- D'Amato CR, Griggs S, McCoy B: Nighttime bracing with the Providence brace in adolescent girls with idiopathic scoliosis. Spine. 2001, 26 (18): 2006-2012. 10.1097/00007632-200109150-00014.View ArticlePubMedGoogle Scholar
- de Mauroy JC, Fender P, Tato B, Lusenti P, Ferracane G: Lyon brace. Stud Health Technol Inform. 2008, 135: 327-340.PubMedGoogle Scholar
- de Mauroy JC, Lecante C, Barral F, Daureu D, Gualerzi S, Gagliano R: The Lyon brace. Disabil Rehabil Assist Technol. 2008, 3 (3): 139-145. 10.1080/17483100801904069.View ArticlePubMedGoogle Scholar
- Emans JB, Kaelin A, Bancel P, Hall JE, Miller ME: The Boston bracing system for idiopathic scoliosis. Follow-up results in 295 patients. Spine. 1986, 11 (8): 792-801. 10.1097/00007632-198610000-00009.View ArticlePubMedGoogle Scholar
- Grivas T, Vasiliadis E, Chatziargiropoulos T, Polyzois V, Gatos K: The effect of a modified Boston brace with anti-rotatory blades on the progression of curves in idiopathic scoliosis: aetiologic implications. Pediatric Rehabilitation. 2003, 6: 237-242.View ArticlePubMedGoogle Scholar
- Veldhuizen AG, Cheung J, Bulthuis GJ, Nijenbanning G: A new orthotic device in the non-operative treatment of idiopathic scoliosis. Med Eng Phys. 2002, 24 (3): 209-218. 10.1016/S1350-4533(02)00008-5.View ArticlePubMedGoogle Scholar
- Blount W, Moe J: The Milwaukee Brace. 1973, Baltimore: The William and Wilkins CompanyGoogle Scholar
- Lonstein JE, Winter RB: The Milwaukee brace for the treatment of adolescent idiopathic scoliosis. A review of one thousand and twenty patients. J Bone Joint Surg Am. 1994, 76 (8): 1207-1221.PubMedGoogle Scholar
- Maruyama T, Takeshita K, Kitagawa T: Milwaukee brace today. Disabil Rehabil Assist Technol. 2008, 3 (3): 136-138. 10.1080/17483100801904036.View ArticlePubMedGoogle Scholar
- The Charleston Bending Brace. An Orthothist's Guide to Scoliosis Management. [http://www.srs.org/professionals/bracing_manuals/section7.pdf]
- Price CT, Scott DS, Reed FR, Sproul JT, Riddick MF: Nighttime bracing for adolescent idiopathic scoliosis with the Charleston Bending Brace: long-term follow-up. J Pediatr Orthop. 1997, 17 (6): 703-707.PubMedGoogle Scholar
- Grivas TB, Rodopoulos GI, Bardakos NV: Night-time braces for treatment of adolescent idiopathic scoliosis. Disabil Rehabil Assist Technol. 2008, 3 (3): 120-129. 10.1080/17483100801903954.View ArticlePubMedGoogle Scholar
- Coillard C, Vachon V, Circo AB, Beausejour M, Rivard CH: Effectiveness of the SpineCor brace based on the new standardized criteria proposed by the scoliosis research society for adolescent idiopathic scoliosis. J Pediatr Orthop. 2007, 27 (4): 375-379. 10.1097/01.bpb.0000271330.64234.db.View ArticlePubMedGoogle Scholar
- Negrini S, Atanasio S, Zaina F, Romano M, Parzini S, Negrini A: End-growth results of bracing and exercises for adolescent idiopathic scoliosis. Prospective worst-case analysis. Stud Health Technol Inform. 2008, 135: 395-408.PubMedGoogle Scholar
- Negrini S, Carabalona R: Social acceptability of treatments for adolescent idiopathic scoliosis: a cross-sectional study. Scoliosis. 2006, 1: 14-10.1186/1748-7161-1-14.View ArticlePubMedPubMed CentralGoogle Scholar
- Negrini S, Fusco C, Romano M, Zaina F, Atanasio S: Clinical and postural behaviour of scoliosis during daily brace weaning hours. Stud Health Technol Inform. 2008, 140: 303-306.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.