Radiation dose of digital radiography (DR) versus micro-dose x-ray (EOS) on patients with adolescent idiopathic scoliosis: 2016 SOSORT- IRSSD “John Sevastic Award” Winner in Imaging Research
© The Author(s). 2017
Received: 10 June 2016
Accepted: 7 December 2016
Published: 29 December 2016
Patients with adolescent idiopathic scoliosis (AIS) frequently receive x-ray imaging at diagnosis and subsequent follow monitoring. The ionizing radiation exposure has accumulated through their development stage and the effect of radiation to this young vulnerable group of patients is uncertain. To achieve the ALARA (as low as reasonably achievable) concept of radiation dose in medical imaging, a slot-scanning x-ray technique by the EOS system has been adopted and the radiation dose using micro-dose protocol was compared with the standard digital radiography on patients with AIS.
Ninety-nine participants with AIS underwent micro-dose EOS and 33 underwent standard digital radiography (DR) for imaging of the whole spine. Entrance-skin dose was measured using thermoluminescent dosimeters (TLD) at three regions (i.e. dorsal sites at the level of sternal notch, nipple line, symphysis pubis). Effective dose and organ dose were calculated by simulation using PCXMC 2.0. Data from two x-ray systems were compared using independent-samples t-test and significance level at 0.05. All TLD measurements were conducted on PA projection only. Image quality was also assessed by two raters using Cobb angle measurement and a set of imaging parameters for optimization purposes.
Entrance-skin dose from micro-dose EOS system was 5.9–27.0 times lower at various regions compared with standard DR. The calculated effective dose was 2.6 ± 0.5 (μSv) and 67.5 ± 23.3 (μSv) from micro-dose and standard DR, respectively. The reduction in the micro-dose was approximately 26 times. Organ doses at thyroid, lung and gonad regions were significantly lower in micro-dose (p < 0.001). Data were further compared within the different gender groups. Females received significantly higher (p < 0.001) organ dose at ovaries compared to the testes in males. Patients with AIS received approximately 16–34 times lesser organ dose from micro-dose x-ray as compared with the standard DR. There was no significant difference in overall rating of imaging quality between EOS and DR. Micro-dose protocol provided enough quality to perform consistent measurement on Cobb angle.
Entrance-skin dose, effective dose and organ dose were significantly reduced in micro-dose x-ray. The effective dose of a single micro-dose x-ray (2.6 μSv) was less than a day of background radiation. As AIS patients require periodic x-ray follow up for surveillance of curve progression, clinical use of micro-dose x-ray system is beneficial for these young patients to reduce the intake of ionizing radiation.
KeywordsAIS Radiation Micro-dose 2D/3D slot-scanning x-ray Entrance skin dose Effective dose Organ dose Thermoluminescent dosimeters
Patients with adolescent idiopathic scoliosis (AIS) suffer from 3-dimensional spinal deformities. The onset and progress occur during their youth stage and usually become stable after skeletal maturity. The current gold standard of diagnosis is made based on the measurement of a Cobb angle larger than 10°. As the chance of curve progression increases in younger patients with greater initial Cobb angle [1, 2], patients receive brace treatment and follow-up monitoring in routine basis at young age. During each follow-up, patients undergo digital radiography to capture images of spine which allow physicians to monitor their curve progression over time. As the treatment of AIS covers a relatively long period during their adolescence, the accumulation of ionizing radiation has become a concern for this vulnerable group of teenagers.
Ionizing radiation from x-ray has high enough energy to break molecular bonding in humans. Damaged bonding repaired incorrectly could affect chromosome to induce cancer [3, 4]. The accumulated ionizing radiation increases the probability of adverse health issues and uncertainties including cancer and abnormal pregnancy to patients. Retrospective studies indicated patients with AIS who frequently received x-ray have approximately 2 and 3% increased lifetime risk of breast cancer and heritable defect, respectively [5–7], and higher risks of unsuccessful attempts at pregnancy, spontaneous abortions, infants with congenital malformations and lower birthweight . As pediatric patients have a longer lifetime to manifest radiation damage than adults and the adverse effects could appear years after exposure, it is important to call for special attention in radiation protection and apply any available methods to achieve the principle of ALARA (as low as reasonably achievable) to minimize the radiation.
Considerable efforts and improvements have been made to reduce the radiation dose from x-ray imaging. Increasingly, conventional film based radiographies are being replaced by the digital ones over the last 15 years. Moreover, the digital technique reduces the number of x-ray retake from 5.5% for conventional to 1.0%, which significantly avoids repeated exposure . Literatures also suggest that the change of image orientation from anterior-posterior (AP) view to posterior-anterior (PA) view could greatly reduce the organ dose by approximately three- to eight-fold to the breasts and thyroid because of the lower sensitive organ dose to the anterior structures; thereby, it is suggested for routine spine examinations [10, 11]. Other technical improvement has been made to reduce the exposure including the use of 3-phase x-ray machines and high-speed x-ray films . A new implementation of radiography, the slot-scanning x-ray by EOS 2D/3D system (EOS Imaging, Paris, France), has been adopted recently and it has a great advantage in capturing x-ray images using very low radiation dose. The EOS system is equipped with two sets of x-ray tube mounted at right angles, a biplanar design, and utilizes the multi-wire proportional chamber (MWPC) to detect charged particles and photons for simultaneous acquisition of frontal and lateral images . The application of EOS mainly involves clinical measurement and analysis of spinal curvature in AIS [14–16], bone fracture , torsion [18–21], orientation and alignment of spine and lower body limb [22–25]. The accuracy, reliability and reproducibility of curve measurement using 3D reconstruction feature in EOS have also been tested and the results are comparable with manual 2D method and CT data [26, 27].
An early experimental study indicated that the entrance skin dose from slot-scanning x-ray technique using MWPC detector was reduced by 13 times at PA orientation and 15 times at lateral in a full spine procedure compared to the conventional film based radiography while no significant loss of diagnostic information . The current EOS system also embedded with MWPC provides two strengths of acquisition protocols e.g. the standard low-dose and the micro-dose. Previous literature reported the entrance skin dose, using low-dose protocol, was reduced by 6 to 9 times with improved image quality compared with computed radiography . A phantom based radiological study reported the effective dose of a full spine examination using EOS low-dose protocol was 290 μSv for an adult and 200 μSv for a child .
As micro-dose is a relatively new protocol from EOS, very limited number of publications is available regarding the radiation dose and image quality. To the best of our knowledge, a recent study reported the radiation exposure was reduced by 5.5 and 45 times compared to the standard low-dose and conventional radiography respectively. However, details on methodology to measure and calculate the air kerma have not been fully presented as air kerma measures the amount of kinetic energy deposited or absorbed in a unit mass of air which is corresponding to the entrance skin dose .
In this study, radiation impact on patients with AIS during whole spine imaging using micro-dose EOS and standard digital radiography (DR) were investigated and compared systematically. Comprehensive measurement of various radiation parameters including entrance skin dose, effective dose and organ dose were included. Entrance skin dose is a direct measurement of radiation output at the point of skin entry for x-ray examinations, and effective dose is a calculated value, commonly in the unit of milli-sivert (mSv) or micro-sivert (μSv), that takes the absorbed dose to all organs of the body, the relative harm level of the radiation and the sensitivities of each organ to radiation into account. Image quality from both techniques was also assessed using criteria for diagnostic radiographic images.
Demographics of patients
EOS micro-dose (n = 99)
Digital radiography (n = 33)
18 male, 81 female
11 male, 22 female
Micro-dose full spine x-ray images were taken from EOS slot-scanning system, newly implemented for radiographic examination, with a total filtration of 0.1 mm copper (Cu) and an x-ray tube anode angle of 7°. Images acquired from digital radiography (Definium 8000, General Electric, United States) with total filtration of 2.7 mm aluminum equivalent employed stitching method to develop a full spine image. All images were taken at PA standing orientation with both arms raised and hands holding the handling bar during the procedure in micro-dose EOS and were protected by collimators in digital radiography.
Measurement of radiation dose
Parameters in EOS micro-dose and standard digital radiography
Beam width (cm)
Beam height (cm)
32.9 (3.04) – in 3 sections
Projection angle (degree)
X-ray tube potential (kv)
X-ray current (mA)
X-ray tube anode angle (degree)
Filter thickness (mm)
Scanning Time (sec/msec)
12.09 (3.92) Sternal notch
33.03 (13.72) Nipple line
53.14 (18.04) Symphysis pubis
Monte Carlo simulation parameters: Maximum energy
Monte Carlo simulation parameters: Number of photons
Standard digital radiography used stitching method to connect three sections of the x-ray into a full spine image. So the simulation in PCXMC was also performed in three sections to calculate the effective dose and organ dose based on DAP and TLD reading at the dorsal sites at the level of sternal notch, nipple line and pubic symphysis level. Scanning range and area of measurement were obtained during the procedure. EOS imaging employed the slot-scanning technique to obtain the spinal images. Continuous scanning allowed one single shot radiation exposure avoiding repeated exposures in duplicated regions, which often happened in standard digital radiography. On each patient, only one simulation, (including the full body), was performed to calculate effective dose and organ dose for EOS micro-dose protocol.
Evaluation of image quality
Images obtained from EOS micro-dose and standard digital radiography were compared using inter-observer variation based on Cobb angle measurement and image quality evaluation according to Kogon et al  and Cook et al . Two raters, who had undergone training to measure Cobb angle using standardized method with over 3 years of experience in AIS related research, performed the rating independently.
Equality of variances was measured by Levene’s Test and equality of means of DAP, entrance skin dose, effective dose, and organ dose were analyzed between group by independent samples t-test using SPSS 20 (SPSS, Chicago, IL). Results were further divided into gender groups (e.g. female in EOS, male in EOS, female in DR and male in DR) and comparisons between different genders were also conducted by independent samples t-test within EOS and digital radiography. Results were presented in mean and standard deviation and statistical significant level was set at p < 0.05. Intra-class correlation coefficient (ICC) was used to measure inter-rater reliability from Cobb angles obtained from two raters. It allowed us to evaluate whether or not image quality from micro-dose x-ray or standard digital radiography would affect raters’ consistency in measuring Cobb angle. Image quality evaluation according to Kogon et al  and Cook et al  allowed the assessment for optimization of images obtained from micro-dose EOS and standard digital radiography. All nine parameters plus the overall rating were compared between EOS and digital radiography using non-parametric Mann–Whitney U test for ordinal data.
Statistical results of radiation dose between EOS and standard digital radiography
EOS (n = 99)
Digital Radiography (n = 33)
Entrance Skin Dose (μGy)
- Sternal Notcha
- Nipple Linea
- Symphysis Pubisa
Effective Dose (μSv)
Organ Dose (μGy)
- Reproductive Organ
Results from EOS micro-dose in different gender group
Female in EOS (n = 81)
Male in EOS (n = 18)
Entrance Skin Dose (μGy)
- Sternal Notcha
- Nipple Linea
- Symphysis Pubisa
Effective Dose (μSv)
Organ Dose (μGy)
- Reproductive Organ
Results from standard digital radiography in different gender group
Female in DR (n = 22)
Male in DR (n = 11)
Entrance Skin Dose (μGy)
- Sternal Notcha
- Nipple Linea
- Symphysis Pubisa
Effective Dose (μSv)
Organ Dose (μGy)
- Reproductive Organ
Average ratings from Rater 1 and 2 on images from EOS and digital radiography
Rater 1 EOS
Rater 1 DR
Rater 2 EOS
Rater 2 DR
Collimation marks seen (0–3)
Presence of holder’s hands (0–3)
Gonad/additional lead protection (0–3)
Entrance skin dose (ESD) was a direct measurement of radiation absorbed by skin. It was measured in the unit of gray (Gy) which one Gy of the dose was equivalent to one joule of energy deposited in a kilogram of matter (J/kg). In this study, it was measured at the back of each subject corresponding to the level of sternal notch, nipple line and pubic symphysis. These three regions were selected due to the relatively high radio-sensitivity of their corresponding tissues/organs (e.g. thyroid, lung, breast, and gonads) . Micro-dose EOS produced consistent air kerma with the slot-scanning technique. ESD exposed on patients was therefore very stable at all three regions at a very low dose as shown in Table 3. In standard digital radiography, ESD varied and the highest dose was measured at pubic symphysis mainly due to the longer exposure time. In both EOS and standard digital radiography, male received significantly higher ESD at dorsal site of sternal notch (p = 0.023 in EOS, p = 0.013 in DR) compared to female, due to larger input current, longer duration of exposure and a larger area of exposure corresponding to their body size. To further conduct the assessment on biological effect of radiation on patients who underwent x-ray, effective dose was calculated using PCXMC simulation.
All effective doses presented in this study were based on the weighting factor in the latest update from ICRP 103 . Effective dose considered the biological effectiveness of different types of tissues. It was defined as the multiplication of equivalent dose to tissue weighting factor at specific organ. Patients who underwent micro-dose EOS were only exposed to about 3.9% of the effective dose in standard DR from a full spine procedure at PA orientation. The reduction was approximately 26 times (2.6 μSv in EOS versus 67.5 μSv in standard DR). Based on in-house data of 513 patients with AIS from the out-patient clinic, the average follow-up duration was 4.6 years at a 6 month interval starting at the age of 13.5 years old. Estimated number of x-ray taken would be 9.2, which meant patients with AIS in average could reduce almost 600 μSv of effective dose from using micro-dose in accumulation from their adolescent stage or equivalent to approximately one third to a quarter of effective dose from natural background radiation in a year . The effective dose of a single micro-dose x-ray (2.6 μSv) was less than a day of background radiation . Specific organs were taken into account due to their greater radio-sensitivity and organ dose at thyroid, lung and reproductive organs were compared.
In organ dose comparison, data were divided into male and female. Organ dose was the absorbed dose averaged over an organ. Results from EOS and standard digital radiography both indicated that female received significantly higher dose at ovaries compared to testes. It could possibly explain why female patients with AIS had higher risks of unsuccessful attempts at pregnancy, spontaneous abortions and abnormalities in infants as suggested by Goldberg et al. . While the accumulated ionizing radiation could possibly induce adverse effects on both patients underwent x-ray and the development of their fetus in long term, micro-dose protocol became exceptionally valuable to the vulnerable group of adolescent especially in females. The lungs in males was exposed to the greatest organ dose due to its high radio-sensitivity and relatively large volume. But no statistically significant difference was observed within gender groups (p = 0.31 in EOS, p = 0.07 in DR). Besides radiation dose, image quality was also assessed.
Radiation measurement and comparison were conducted on PA plane as the TLD measurement will be difficult to interpret if both PA and lateral views were obtained by the biplanar EOS system due to contamination of the TLD reading by x-ray sources from two different directions. For the patients, age (p = 0.01) and Cobb angle (p = 0.02) were statistically significant different between the two groups as shown in Table 1. Age could be a potential factor that might affect the effective dose simulation but it was mainly used to estimate the risk of death due to radiation-induced cancer which was not included in this study. There were no significant differences in height (p = 0.96) and weight (p = 0.22) between the two groups indicated the exposure area would be similar. Also, no significant difference was observed in bone maturity as the Risser sign was similar between groups (p = 0.70), indicating an absence of difference in biological response of bone tissues. Cobb angle was not strictly controlled as the main purpose of this study was not investigating the etiology of AIS but to measure the radiation exposure. It was not a parameter for the simulation of effective dose and has no known relation to the radiation, so its effect should be minimal. The unequal sample size would be a concern because majority of the follow up AIS patients underwent micro-dose protocol after the implementation of the EOS machine except for those who could not stand firm and steady underwent digital radiography. The number of AIS patients from digital radiography was limited.
We concluded that micro-dose EOS provided comparable and clinically useful images of the whole spine for AIS patients while significantly reduced radiation exposure. We suggest that patients with AIS undergo initial x-ray with standard digital radiography to eliminate differential diagnosis and micro-dose EOS for follow-up purpose, given that no suspicion of bone metastasis, fracture or other complaints existed, to reduce the accumulation of ionizing radiation in long term.
Adolescent idiopathic scoliosis
As low as reasonably achievable
Intra-class correlation coefficient
Multi-wire proportional chamber
We would like to thank our physicists Dr. Louis Lee, Ms. Lee Wai-yee and Mr. Lam Chi-ho from the Division of Medical Physics, Department of Clinical Oncology at the Prince of Wales Hospital for providing their professional advice and service on collecting data from thermoluminescent dosimeters (TLD).
The purposed study was partially supported by the SH Ho Scoliosis Research Laboratory (7104486) and Kai Cheong Tong for equipment funding.
Availability of data and materials
The dataset supporting the conclusions of this article has been made available.
SH – analysis and interpretation of data, drafting of the manuscript. JP –interpretation of data, drafting of the manuscript. JW – acquisition of data. TL – clinical diagnosis and subject referral. BN – clinical diagnosis and subject referral. JC – conception and design, critical revision of the manuscript. WC – conception and design, critical revision of the manuscript. All authors read and approved the final manuscript.
The authors declare that they have no competing interests.
Consent for publication
Consent to publish was obtained from all volunteers and their parent or legal guardian.
Ethics approval and consent to participate
The research protocol was approved by the Clinical Research Ethics Committee of the institution and conducted in compliance with the principles of Declaration of Helsinki. Written informed consents were obtained from both volunteers and their parent (or legal guardian).
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Weinstein SL, Dolan LA, Cheng JC, Danielsson A, Morcuende JA. Adolescent idiopathic scoliosis. Lancet. 2008;371:1527–37.View ArticlePubMedGoogle Scholar
- Deng M, Hui SC, Yu FW, Lam TP, Qiu Y, Ng BK, et al. MRI-based morphological evidence of spinal cord tethering predicts curve progression in adolescent idiopathic scoliosis. Spine J. 2015;15:1391–401.View ArticlePubMedGoogle Scholar
- Shah DJ, Sachs RK, Wilson DJ. Radiation-induced cancer: a modern view. Br J Radiol. 2012;85:e1166–73.View ArticlePubMedPubMed CentralGoogle Scholar
- Thompson LH. Recognition, signaling, and repair of DNA double-strand breaks produced by ionizing radiation in mammalian cells: the molecular choreography. Mutat Res. 2012;751:158–246.View ArticlePubMedGoogle Scholar
- Levy AR, Goldberg MS, Hanley JA, Mayo NE, Poitras B. Projecting the lifetime risk of cancer from exposure to diagnostic ionizing radiation for adolescent idiopathic scoliosis. Health Phys. 1994;66:621–33.View ArticlePubMedGoogle Scholar
- Doody MM, Lonstein JE, Stovall M, Hacker DG, Luckyanov N, Land CE. Breast cancer mortality after diagnostic radiography: findings from the U.S. Scoliosis Cohort Study. Spine (Phila Pa 1976). 2000;25:2052–63.View ArticleGoogle Scholar
- Bone CM, Hsieh GH. The risk of carcinogenesis from radiographs to pediatric orthopaedic patients. J Pediatr Orthop. 2000;20:251–4.PubMedGoogle Scholar
- Goldberg MS, Mayo NE, Levy AR, Scott SC, Poitras B. Adverse reproductive outcomes among women exposed to low levels of ionizing radiation from diagnostic radiography for adolescent idiopathic scoliosis. Epidemiology. 1998;9:271–8.View ArticlePubMedGoogle Scholar
- Akhtar W, Aslam M, Ali A, Mirza K, Ahmad N. Film retakes in digital and conventional radiography. J Coll Physicians Surg Pak. 2008;18:151–3.PubMedGoogle Scholar
- Luo TD, Stans AA, Schueler BA, Larson AN. Cumulative Radiation Exposure With EOS Imaging Compared With Standard Spine Radiographs. Spine Deformity. 2015;3:144–50.View ArticlePubMedGoogle Scholar
- Levy AR, Goldberg MS, Mayo NE, Hanley JA, Poitras B. Reducing the lifetime risk of cancer from spinal radiographs among people with adolescent idiopathic scoliosis. Spine (Phila Pa 1976). 1996;21:1540–7. discussion 8.View ArticleGoogle Scholar
- Knott P, Pappo E, Cameron M, Demauroy J, Rivard C, Kotwicki T, et al. SOSORT 2012 consensus paper: reducing x-ray exposure in pediatric patients with scoliosis. Scoliosis. 2014;9:4.View ArticlePubMedPubMed CentralGoogle Scholar
- Charpak G, Bouclier R, Bressani T, Favier J, Zupancic C. Use of Multiwire Proportional Counters to Select and Localize Charged Particles. Nucl Instrum Methods. 1968;62:262.View ArticleGoogle Scholar
- Amzallag-Bellenger E, Uyttenhove F, Nectoux E, Moraux A, Bigot J, Herbaux B, et al. Idiopathic scoliosis in children and adolescents: assessment with a biplanar X-ray device. Insights Imaging. 2014;5:571–83.View ArticlePubMedPubMed CentralGoogle Scholar
- Begon M, Scherrer SA, Coillard C, Rivard CH, Allard P. Three-dimensional vertebral wedging and pelvic asymmetries in the early stages of adolescent idiopathic scoliosis. Spine J. 2015;15:477–86.View ArticlePubMedGoogle Scholar
- Courvoisier A, Drevelle X, Vialle R, Dubousset J, Skalli W. 3D analysis of brace treatment in idiopathic scoliosis. Eur Spine J. 2013;22:2449–55.View ArticlePubMedPubMed CentralGoogle Scholar
- Briot K, Fechtenbaum J, Etcheto A, Kolta S, Feydy A, Roux C. Diagnosis of vertebral fractures using a low-dose biplanar imaging system. Osteoporos Int. 2015;26:2649–55.View ArticlePubMedGoogle Scholar
- Buck FM, Guggenberger R, Koch PP, Pfirrmann CW. Femoral and tibial torsion measurements with 3D models based on low-dose biplanar radiographs in comparison with standard CT measurements. AJR Am J Roentgenol. 2012;199:W607–12.View ArticlePubMedGoogle Scholar
- Rosskopf AB, Ramseier LE, Sutter R, Pfirrmann CW, Buck FM. Femoral and tibial torsion measurement in children and adolescents: comparison of 3D models based on low-dose biplanar radiography and low-dose CT. AJR Am J Roentgenol. 2014;202:W285–91.View ArticlePubMedGoogle Scholar
- Kadoury S, Shen J, Parent S. Global geometric torsion estimation in adolescent idiopathic scoliosis. Med Biol Eng Comput. 2014;52:309–19.View ArticlePubMedGoogle Scholar
- Meyrignac O, Moreno R, Baunin C, Vial J, Accadbled F, Sommet A, et al. Low-dose biplanar radiography can be used in children and adolescents to accurately assess femoral and tibial torsion and greatly reduce irradiation. Eur Radiol. 2015;25:1752–60.View ArticlePubMedGoogle Scholar
- Ilharreborde B, Vidal C, Skalli W, Mazda K. Sagittal alignment of the cervical spine in adolescent idiopathic scoliosis treated by posteromedial translation. Eur Spine J. 2013;22:330–7.View ArticlePubMedGoogle Scholar
- Sutter R, Pfirrmann CW, Espinosa N, Buck FM. Three-dimensional hindfoot alignment measurements based on biplanar radiographs: comparison with standard radiographic measurements. Skeletal Radiol. 2013;42:493–8.View ArticlePubMedGoogle Scholar
- Morvan G, Mathieu P, Vuillemin V, Guerini H, Bossard P, Zeitoun F, et al. Standardized way for imaging of the sagittal spinal balance. Eur Spine J. 2011;20 Suppl 5:602–8.View ArticlePubMedPubMed CentralGoogle Scholar
- Lazennec JY, Rousseau MA, Rangel A, Gorin M, Belicourt C, Brusson A, et al. Pelvis and total hip arthroplasty acetabular component orientations in sitting and standing positions: measurements reproductibility with EOS imaging system versus conventional radiographies. Orthop Traumatol Surg Res. 2011;97:373–80.View ArticlePubMedGoogle Scholar
- Somoskeoy S, Tunyogi-Csapo M, Bogyo C, Illes T. Accuracy and reliability of coronal and sagittal spinal curvature data based on patient-specific three-dimensional models created by the EOS 2D/3D imaging system. Spine J. 2012;12:1052–9.View ArticlePubMedGoogle Scholar
- Glaser DA, Doan J, Newton PO. Comparison of 3-dimensional spinal reconstruction accuracy: biplanar radiographs with EOS versus computed tomography. Spine (Phila Pa 1976). 2012;37:1391–7.View ArticleGoogle Scholar
- Kalifa G, Charpak Y, Maccia C, Fery-Lemonnier E, Bloch J, Boussard JM, et al. Evaluation of a new low-dose digital x-ray device: first dosimetric and clinical results in children. Pediatr Radiol. 1998;28:557–61.View ArticlePubMedGoogle Scholar
- Deschenes S, Charron G, Beaudoin G, Labelle H, Dubois J, Miron MC, et al. Diagnostic imaging of spinal deformities: reducing patients radiation dose with a new slot-scanning X-ray imager. Spine (Phila Pa 1976). 2010;35:989–94.View ArticleGoogle Scholar
- Damet J, Fournier P, Monnin P, Sans-Merce M, Ceroni D, Zand T, et al. Occupational and patient exposure as well as image quality for full spine examinations with the EOS imaging system. Med Phys. 2014;41(6):063901. http://onlinelibrary.wiley.com/wol1/doi/10.1118/1.4873333/full.
- Ilharreborde B, Ferrero E, Alison M, Mazda K. EOS microdose protocol for the radiological follow-up of adolescent idiopathic scoliosis. Eur Spine J. 2016;25:526–31.View ArticlePubMedGoogle Scholar
- Servomaa A, Tapiovaara M. Organ Dose Calculation in Medical X Ray Examinations by the Program PCXMC. Radiat Prot Dosim. 1998;80:213–9.View ArticleGoogle Scholar
- Kogon PL, Lumsden R. How do you critique your radiographs? J Can Chiropr Assoc. 1993;37:230–2.PubMed CentralGoogle Scholar
- Cook JV, Kyriou JC, Pettet A, Fitzgerald MC, Shah K, Pablot SM. Key factors in the optimization of paediatric X-ray practice. Br J Radiol. 2001;74:1032–40.View ArticlePubMedGoogle Scholar
- The 2007 Recommendations of the International Commission on Radiological Protection. ICRP publication 103. Ann ICRP. 2007; 37:1-332.
- Canadian Nuclear Safety Commission. Fact sheet – Natural background radiation. 2014-11-19 2013. http://nuclearsafety.gc.ca/eng/pdfs/Fact_Sheets/Fact-Sheet-Background-Radiation-eng.pdf. Accessed Nov 2014.