- Open Access
Measurement of axial vertebral rotation using three-dimensional ultrasound images
© Vo et al; licensee BioMed Central Ltd. 2015
- Published: 11 February 2015
Axial vertebral rotation (AVR) is one of the important parameters to evaluate the severity and predict the progression of scoliosis. However, the AVR measurements on radiographs may underestimate its actual value. This pilot study investigated a new three-dimensional (3D) ultrasound method to measure AVR.
Three cadaveric vertebrae T7, L1, and L3 were scanned with a 3D medical ultrasound system. Nine sets of ultrasound data, the vertebral rotation from 0 to 40° with 5° increments, were recorded from each vertebra. An in-house program was developed to reconstruct and measure the 3D vertebral images. The rotation of each reconstructed vertebra was determined by the angle between the line going through the centres of either laminae (L-L) or transverse processes (TP-TP) and a reference vertical plane. Three raters measured the rotation in 3 sessions, in which they used the mouse pointer to select the L-L or TP-TP according to their knowledge of vertebral anatomy. The program detected the 3D coordinates of these points and calculated the AVR.
The intra-class correlation coefficients (ICCs) were used to calculate the intra-reliability and inter-reliability. The mean absolute difference (MAD±SD) and the range of difference (RD) between the actual values and the average measurements of each rater were computed to evaluate the accuracy of methods.
When rotation was greater than 30° for both L1 and L3, all raters found it difficult to determine one of the lamina areas due to the lack of ultrasound information in an area behind the spinous process. Therefore, the corresponding measurements were excluded. The ICC values of the intra-reliability (L-L, TP-TP) for the three raters were (0.987, 0.991), (0.989, 0.998) and (0.997, 1.000), respectively; meanwhile, the inter-reliability were 0.991 for (L-L) and 0.992 for (TP-TP). All ICC values were greater than 0.98 indicating both methods were highly reliable. The MAD±SD values (L-L, TP-TP) for the three raters were (1.5±0.3°, 1.2±0.2°), (1.6±0.3°, 1.3±0.3°), and (1.7±0.5°, 0.9±0.2°), respectively. The RD (L-L, TP-TP) were (0-4.5°, 0-3.5°), (0-5.1°, 0-4.3°), and (0-5.1°, 0-2.8°) for the three raters, respectively.
The (L-L) and (TP-TP) methods could be used to measure AVR reliability from the 3D ultrasound images.
- Adolescent Idiopathic Scoliosis
- Cobb Angle
- Spinous Process
- Transverse Process
- Mean Absolute Difference
Scoliosis is a complex 3D deformity of spine associated with axial vertebral rotation. Adolescent Idiopathic Scoliosis (AIS) accounts for about 80% of all cases and affects 1.5 – 3% of the population . The Cobb angle and axial vertebral rotation are two major factors which can be used to measure the severity of scoliosis, estimate the risk of progression and evaluate the treatment outcomes. Currently, the Cobb angle is gold standard recommended by the Scoliosis Research Society. However, the AVR measurement is often ignored in scoliosis clinics due to time consumption.
Ultrasound has been proposed as an alternative to X-ray based approaches to monitor and image AIS [2–5]. Spinous processes, laminae, transverse processes can be visualized and used as landmarks to measure the Cobb angle and the AVR [2–5] on the frontal or transverse planes of the ultrasound images.
The objectives of this pilot study were to investigate the reliability and the accuracy of the AVR measurements based on either the centres of laminae or the centres of transverse processes on 3D ultrasound images. The ethical approval for this study was granted by the local health research ethics board.
Materials and experimental setup
Three cadaveric vertebrae T7, L1, and L3 were scanned to acquire 27 data sets. An in-house developed program was then used to reconstruct and measure the 3D vertebral images.
The AVR measurement
Three raters who were blinded with the rotation information measured the rotation in 3 sessions in one week apart to minimize memory bias. The intra-class correlation coefficients (ICCs), two-way random and absolute agreement, were used to calculate the intra-reliability and inter-reliability. The accuracy of the measurement method was determined based on the mean absolute difference (MAD ± SD) and the range of difference (RD) between the true values and the average measurements of each rater.
The intra-reliability among three measurements of each rater.
L – L
TP - TP
The mean absolute difference [absolute(mean measured value – actual value)].
L – L
1.5 ± 0.3°
1.6 ± 0.3°
1.7 ± 0.5°
TP - TP
1.2 ± 0.2°
1.3 ± 0.3°
0.9 ± 0.2°
The range of absolute difference [absolute(mean measured value – actual value)]
L – L
0 - 4.5°
0 - 5.1°
0 - 5.1°
TP - TP
0 - 3.5°
0 - 4.3°
0 - 2.8°
The mean absolute difference between two raters
Rater 1 vs. Rater 2
Rater 2 vs. Rater 3
Rater 1 vs. Rater 3
L – L
1.06 ± 0.93°
1.18 ± 1.59°
0.93 ± 1.42°
TP - TP
1.34 ± 1.32°
0.94 ± 0.68°
0.71 ± 0.85°
Three different cadaveric vertebrae with different structures were used to investigate the feasibility of measuring the AVR from the 3D reconstructed ultrasound images. The ICC values of the intra-reliability and the inter-reliability were greater than 0.98 indicating both methods (L-L and TP-TP) were highly reliable. The ICC, the MAD and RD values of the TP-TP method are higher or better than the L-L method; however, it does not show a clinical significant difference (3-5o error) on the measurements. Both the L-L and TP-TP methods could measure the AVR reliability from the 3D ultrasound images. In in-vivo cases, the L-L method is preferable as the transverse processes may be interfered with ribs. Also, the distance between the laminae is shorter, which is more likely to be covered by the ultrasound signals.
Due to the ultrasound characteristics, the more the vertebra rotates, the higher chance the spinous process blocks the ultrasound signals which may cause missing lamina data.
The results demonstrated that L-L and TP-TP could be used interchangeably to measure AVR from the 3D ultrasound images. The actual accuracy and rotation measurement limitation may require more investigation in in-vivo cases.
This is the extended abstract of IRSSD 2014 program book .
We would like to acknowledge Vietnam International Education Development (VIED), Edmonton Orthopaedic Research Committee, Women and Children’s Health Research Institute (WCHRI), and Natural Sciences and Engineering Research Council of Canada (NSERC) for their financial supports. We also would like to thank Ms. Amanda Chan and Ms. Michelle Young for their technical supports.
Publication charges for this collection were funded by IRSSD 2014 Sapporo.
This article has been published as part of Scoliosis Volume 10 Supplement 2, 2015: Research into Spinal Deformities 9: Short Papers from the IRSSD 2014 Meeting. The full contents of the supplement are available online at http://www.scoliosisjournal.com/supplements/10/S2.
- Lonstein JE: Adolescent Idiopathic Scoliosis. Lancet. 1994, 344: 1407-1412. 10.1016/S0140-6736(94)90572-X.View ArticlePubMedGoogle Scholar
- Suzuki S, Yamamuro T, Shikata J, Shimizu K, Iida H: Ultrasound measurement of vertebral rotation in idiopathic scoliosis. J Bone Joint Surg Br. 1989, 71 (2): 252-255.PubMedGoogle Scholar
- Chen W, Lou EHM, Le LH: Using ultrasound imaging to identify landmarks in vertebra models to assess spinal deformity. Proceedings of 33rd Annual International Conference of the IEEE EMBS: August 30 – September 3. 2011, BostonGoogle Scholar
- Purnama KE, Wilkinson MH, Veldhuizen AG, van Ooijen PM, Lubbers J, Burgerhof JG, Sardjono TA, Verkerke GJ: A framework for human spine imaging using a freehand 3D ultrasound system. Technol Health Care. 2010, 18 (1): 1-17.PubMedGoogle Scholar
- Cheung CWJ, Zheng Y: Development of 3-D Ultrasound System for Assessment of Adolescent Idiopathic Scoliosis (AIS). Proceedings of the 6th World Congress of Biomechanics. (WCB 2010) IFMBE; Singapore. Edited by: Lim CT and Goh JCH. 2010, Springer, 584-587.Google Scholar
- Quang Vo, et al: Measurement of Vertebral Rotation Using A Three-dimensional Ultrasound Image. Scoliosis. 2015, 10 (Suppl 1): O40-View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.