- Oral presentation
- Open Access
The effectivity of the new improved Maastricht brace compared with the Boston brace
© Hermus et al; licensee BioMed Central Ltd. 2012
- Published: 27 January 2012
- Public Health
- High Pressure
- Pressure Measurement
- Optimal Result
- Family Support
The brace 2000 was developed to improve the compliance of brace treatment and therefore its results. We further developed the brace 2000 in the Maastricht brace (M brace) to enhance the effectivity of the brace and its wearing comfort.
The mean primary right thoracic curve was 29° uncorrected; the mean secondary curve measured 19°. In the Boston brace group the mean primary right thoracic curve was 22° ; the mean secondary curve measured 16°. The mean corrective force over the lumbar brace pad in standing position was 382 N; over the thoracic brace pad it was 285 N. In the M brace group the mean primary right thoracic curve was 20° ; the mean secondary curve measured 13°. The mean corrective force over the lumbar brace pad in standing position was 373 N; over the thoracic brace pad it was 311 N. Difference in M-brace and Boston brace was not significant, for the lumbar brace pad: 0.12, for the thoracic brace pad 0.07 with ANOVA analysis.
There is a tendency that the M Brace give a higher pressure comparing to the classic Boston brace with a better correction on the spinal radiograph. Bracing results are directly related to compliance with brace treatment; therefore, optimal results cannot be achieved without the patient's cooperation and family support. Brace mechnisms can be explained and understood by giving more insight about pressure/force measurements in the brace. The Quality of life measured with the SRS 22 and Brace questionnaire was higher in the Brace 2000 than the Boston brace.
- Parent S, Newton PO, Wenger DR: Adolescent idiopathic scoliosis: natural history and long-term treatment effects. Instr Course Lect. 2005, 54: 529-36. ReviewPubMedGoogle Scholar
- van den Hout JA, van Rhijn LW, van den Munckhof RJ, van Ooy A: Interface corrective force measurements in Boston brace treatment. Eur. Spine J. 2002, 11 (4): 332-5. 10.1007/s00586-001-0379-1.PubMed CentralView ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an open access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.