This article has Open Peer Review reports available.
A review of the trunk surface metrics used as Scoliosis and other deformities evaluation indices
© Patias et al; licensee BioMed Central Ltd. 2010
Received: 3 May 2010
Accepted: 29 June 2010
Published: 29 June 2010
Although scoliosis is characterized by lateral deviation of the spine, a 3D deformation actually is responsible for geometric and morphologic changes in the trunk and rib cage. In a vast related medical literature, one can find quite a few scoliosis evaluation indices, which are based on back surface data and are generally measured along three planes. Regardless the large number of such indices, the literature is lacking a coherent presentation of the underlying metrics, the involved anatomic surface landmarks, the definition of planes and the definition of the related body axes. In addition, the long list of proposed scoliotic indices is rarely presented in cross-reference to each other. This creates a possibility of misunderstandings and sometimes irrational or even wrong use of these indices by the medical society.
Materials and methods
It is hoped that the current work contributes in clearing up the issue and gives rise to innovative ideas on how to assess the surface metrics in scoliosis. In particular, this paper presents a thorough study on the scoliosis evaluation indices, proposed by the medical society.
More specifically, the referred indices are classified, according to the type of asymmetry they measure, according to the plane they refer to, according to the importance, and relevance or the level of scientific consensus they enjoy.
Surface metrics have very little correlation to Cobb angle measurements. Indices measured on different planes do not correlate to each other. Different indices exhibit quite diverging characteristics in terms of observer-induced errors, accuracy, sensitivity and specificity. Complicated positioning of the patient and ambiguous anatomical landmarks are the major error sources, which cause observer variations. Principles that should be followed when an index is proposed are presented.
Our interest in the study of the trunk surface (TS) deformity is recently increased due to a variety of reasons.
The cosmetic improvement of the trunk after any treatment is of paramount importance to the child under treatment and his family. The TS symmetry is what it is seen and praised by them and not the radiograph itself which is traditionally used by the physician. TS symmetry is also one of the elements intergrading and improving the quality of life of patients, an issue vital for any human being . This was actually the motivation behind both the development of a variety of devices for documentation and evaluation of TS shape and the creation of a variety of indices that are currently used to access the state of such deformities.
The concept is how to collect data related to TS on physiology, to document the pathology, to assess the effect on the TS deformity of any surgical or conservative treatment comparing the pro- to post-treatment state. The characterization of the threshold of normality to pathology is a complex issue that also needs investigation. Although not yet sensitive enough to detect small changes for monitoring of curve natural progression, TS analysis can help to document the external asymmetry associated with different types of spinal curves in scoliosis as well as the cosmetic improvement obtained after surgical interventions .
The review and the evaluation of the TS metrics used as Scoliosis or any deformity evaluation indices would be very useful and would offer some objective assessing tools for the interested physicians.
Scoliosis screening practice
Scoliosis is a deformity of the spine in which there are one or more lateral curvatures deviating from the midline in the coronal plane. Although scoliosis is characterized by lateral deviation of the spine, a 3D deformation actually is responsible for geometric and morphologic changes in the trunk and rib cage .
The goal of scoliosis screening is to detect scoliosis at an early stage, when the deformity is likely to go unnoticed and there is an opportunity for a less invasive method of treatment, or less surgery, than would otherwise be the case. What in reality scoliosis school screening program does, using the scoliometer or any other surface measuring device, is reveal children with surface, mainly thoracic, deformity. It does not reveal the scoliosis per se. It is now definitely accepted that the surface deformity does not accurately predict the magnitude of scoliosis, especially in younger children. As Bunnell characteristically states  "it has become apparent from many reports that, although there is a significant correlation between clinical deformity and radiographic measurement, the standard deviation is so high that it is not possible to reliably predict the degree of curvature from surface topography in any given patient by any technique".
Traditionally, scoliosis screening is done either by Adam test or using other optical techniques, while the radiographic measurement of Cobb angle is considered the golden standard.
The Adam test
The first step in the scoliosis examination is simple inspection. This includes inspection of a standing patient from behind and optical evaluation of asymmetries in shoulders, scapulae, waistline and the distance of the arms from the trunk, as well as the "balance" of the head.
It is reported that Adams test actually demonstrates the rotational component of scoliosis, since the rib prominence is the result of the ribcage rotating along with the spine . The Adams test is considered a very sensitive clinical examination as compared to Cobb angle . However, the sensitivity and specificity1 varies depending upon the skills of the examiner, the location of the curve, and the magnitude of the curve . The range of sensitivity and specificity of the forward bend test varying degrees of scoliosis have been reported as follows [10, 9]:
▪ Thoracic scoliosis with Cobb angle ≥10° - sensitivity 74% - 84%, specificity 78%-93%
▪ Thoracic scoliosis with Cobb angle ≥20° - sensitivity: 92% - 100%, specificity 60% - 91%
▪ Lumbar scoliosis with Cobb angle ≥20° - sensitivity 73%, specificity: 68%
▪ Scoliosis with Cobb angle ≥40° - sensitivity 83%, specificity 99%
High Sensitivity means low rate of false negatives, ie. the number of scoliotic patients classified as normal is small.
High Specificity means low rate of false positives, ie. the number of normal patients classified as scoliotic is small.
It is very important to note that in younger children the concordance of the surface and spinal deformity is weak and it becomes stronger as the children are growing up. Therefore, in younger children with surface trunk asymmetry, the prediction of the spinal deformity alone from the surface topography is inaccurate, simply because surface topography reveals the thoracic cage and the spinal deformity together.
It has also been reported that, in typical screening settings where the prevalence and positive predictive value are relatively low, for every curve >10° detected, there are 1-5 false-positives; similarly, for every curve > 20° detected, there are 3-24 false-positives .
The angle measured by a scoliometer does not correspond to the Cobb angle measured on a radiograph . Furthermore the Cobb angle alone cannot explain the whole of the surface deformity . As a consequence, not all patients with radiographic scoliosis have rotation of the trunk, and not all patients with trunk rotation have radiographic scoliosis . Goldberg  and Kotwicki  agree that "surface parameters corresponding with radiological ones are neither possible nor expedient as both methods focus on different aspects of the deformity. The 3D presentation accompanied by numerical data that is produced in surface topography offers a more complete perspective of the deformity of the back surface and enables a more thorough analysis of the patient's deformity pattern".
The Cobb angle
The degree of curvature in the coronal plane is radiographically measured according to the method of Cobb . The Cobb angle, which is considered the golden standard, is the angle between lines drawn along the upper end plate of the most tilted vertebrae above the curve's apex and the lower end plate of the most tilted vertebrae below the apex. While Cobb angle is the accepted standard for measuring scoliosis on radiographs [17, 18], it has some important limitations [10, 18]:
▪ The Cobb angle describes only one plane of the 3D deformity.
▪ The Cobb angle is not linearly proportional to the severity of scoliosis in a linear fashion (ie, a curve with a Cobb angle of 40° is more than twice as severe as a curve with a Cobb angle of 20°).
▪ Cobb angle measurement has a reported intra-observer variability of 2.8°-4.9° and an inter-observer variability of 6.3°-7.2° [19, 20] when traditional techniques are used. Recent advances in measurements on digitally acquired radiographs provide far more accurate results, with a reported intra-observer and inter-observer variability of 1.3° .
Back surface mapping for scoliosis screening has been used for many years as a valid alternative to either use of x-rays or scoliometer measurements. From the beginning it became clear that "Because surgeons are so familiar with Cobb angle measurements on radiograph, the introduction of new surface shape measures whose meaning may not be readily apparent to clinicians has been difficult" . This explains the effort over the years to relate surface shape parameters with Cobb angle [eg. [23–25, 71]].
However, over the years it became apparent that the Cobb angle measures only one aspect of the 3D deformity and that the correlation between the Cobb angle and the surface parameters is negligible [26, 6, 27]. However, it is noted that the more severe the Cobb angle the more the surface deformity is pronounced.
Lately, many researchers are seriously questioning the effectiveness of such efforts and strong statements, like this appear "Searching for relationship between radiological Cobb angle and surface parameters with making presumption that the higher correlation with Cobb angle, the better the surface technique may be one of the reasons that introduced the surface topography in a blind alley. In fact, Cobb angle is nothing more than a shadow of two limit vertebrae. It is not clear what would be the rationale to expect that so constructed angle should highly correlate with any of the surface describing parameters." . And as Kotwicki states it "When debating on the role of the surface topography in the evaluation of the body morphology in children with idiopathic scoliosis, one should begin with rejecting the dogma of the radiological Cobb angle, as the only gold standard for scoliosis evaluation. " .
Optical systems have been developed as non-invasive imaging techniques. Examples of such systems are the Moiré-fringe mapping , the structured light techniques like the Integrated Shape Imaging System (ISIS) [30–34], or the Quantec system [35, 14, 36, 37] or the Ortelius  scanners, and devices that scan 360° torso profiles [38–41], ultrasound systems , 3D body scanners (eg. Inspeck, Cyberware, TC2, Minolta Vivid, Vitus 3D, etc) , the Formetric video-raster-stereography system http://www.diers.de [72–76] and last but not least stereo-photogrammetric systems [43–47].
Regarding moiré topography, since Takasaki  first introduced it, many other researchers [48–52] have effectively used this technique. Regarding Moiré the following conclusions are useful to our discussion :
▪ There is no correlation between Moiré asymmetry and the Cobb angle
▪ The risk of obtaining false negatives is low (ie. high sensitivity)
▪ The risk of obtaining false positives is high (ie. low specificity)
Metrics in scoliosis evaluation
In a vast related medical literature, one can find quite a few scoliosis evaluation indices, which are based on back surface data and are generally measured along the three planes (coronal, transverse and sagittal). However, there exist no coherent presentation of the underlying metrics, the involved anatomic surface landmarks and the definition of the planes and the related body axes they refer to.
Generally speaking, the scoliosis parameters which have been used up to now belong to one of the following groups: (a) the first group includes indices which are specific to the measurement technique. These indices depend on the measurement technique, which means that cannot be measured and by other means. Such examples are eg. the angles q1 and q2 in QSIS which are angles formed by the tangents to the corresponding fringes in the Moiré system. Obviously, theses cannot me measured with other means than moiré. (b) The second group are indices independent of the measuring technique. This makes them more useful, since they can be used to evaluate scoliosis given that the back surface topography is known in 3D, regardless of the measuring techniques used. Such example is eg. the Angle of Trunk Rotation index, which can be evaluated by scoliometer measurement, by moiré techniques or by any other 3D surface measurement.
Scoliosis surface parameters after 6th SOSORT consensus paper
Position/view of the patient for surface topography measurement [table eighteen]
Position: standing upright
Anatomic surface landmarks to be taken into consideration systematically [table nineteen]
Posterior iliac spines
Surface parameters recommended for systematic use [table twenty]
Body axis definition
Analogous to radiological VCSL
Frontal plane analysis
Sagittal plane analysis
Relation of C7 to S1
Transverse plane analysis
Trunk rotation main curve
Trunk rotation Compensatory curves
Scoliometer ATR measure for transverse plane deformity
Cobb angle measurement as radiological parameter
The reference planes
Anatomic surface landmarks
In order for any measurement taken at different times to be mutually comparable, either the involved metrics should be coordinate-free or they should refer to the same coordinate system.
The first case is rather rare and refers to metrics like areas, volumes, etc. The second case is the usual case and mainly refers to coordinates, angles, distances and the like. In this latter case there is a need to establish a coordinate system, which is stable between the screening sessions.
Body coordinate system
The VCSL (Vertical Central Sacral Line) line is also used in QSIS system, in POTSI and DAPI index definition, etc., while the Z axis definition is compatible to that used in SHS and DAPI (see section 6).
Scoliosis deformity indices: A literature survey
One can find quite a number of scoliotic indices in the literature. Here, for methodological reasons, we are going to present them grouped by the plane they refer to. The reason for such a presentation is twofold: first, to present them in a logical way according to the type of deformity they are able to measure; and secondly to lead the discussion to the degree of correlation existing among them.
Deformity indices measured on the Coronal plane
The spinous process line of Jaremko  and the similar but qualitative indices used in WRVAS (Walter-Reed Visual Assessment Scale) [27, 24] belong to this logic line. Similar to them are also the ASY1 index of  as well as the Integrated Shape Imaging System (ISIS2) LA (Lateral Asymmetry) index . In the latter, a 5th order polynomial is fitted through the spinous process line (as depicted by 19 transversal sections).
Deformity indices measured on the Transverse plane
Deformity indices measured on the Sagittal plane
Discussion and Conclusions
Understanding scoliosis or other trunk deformity is a complex issue since it evolves in three dimensional space. Many technologies have been developed and used over the years and each technology offers new approaches in understanding and describing scoliosis through different sets of indices. Out of this massive data the scientific society has to choose measures and define methodologies in order to optimally diagnose, quantify, document and assess the progression of scoliosis for both clinical treatment and cosmetic improvement.
It is clear now that surface metrics have very little correlation to Cobb angle measurements (eg.  regarding POTSI index). In addition, it has also been reported that patients with double curves have significantly less trunk deformity in both the transverse and coronal plane than patients with thoracic and thoraco-lumbar curves of similar Cobb size .
It should also be clear that indices measured on different planes do not correlate to each other. Examples are Cobb angle vs. Scoliometer angle, Cobb vs. Rib and Flank prominence, etc.
Different indices exhibit quite diverging characteristics in terms of observer-induced errors, accuracy, sensitivity and specificity. Although a complete comparison can not be found in the literature, tabularizing the results and conclusions given by different researchers [56, 58, 59], we give below (Table 2) the specifics for different popular indices.
Characteristics (observer-induced errors, accuracy, sensitivity and specificity) of different popular indices
Threshold for scoliosis cases
Threshold for change (as suggested by Asher)
It is clear that complicated positioning of the patient and ambiguous anatomical landmarks are the major error sources, which cause observer variations. For instance, moiré techniques generally suffer from errors due to malpositions of the patient and generally require strict and cumbersome protocols for positioning the patient. "A major drawback of moiré topography is that while the shape information is displayed, it is not in a form which can be unambiguously interpreted" . POTSI index is reported  to introduce errors due to the difficulty in situating the points involved for calculating the index, as some of them are located in the shaded areas, while they are not anatomical points easily and uniquely identifiable. "The ISIS system lacked accuracy mainly because of the difficulty of distinguishing adequate landmarks due to shadowing effect" .
Therefore, based on the experience gained from this extended literature review, we think it is useful to lay down the principles that should be followed when an index is proposed.
Principles for optimally designed scoliosis Indices
Indices should be measured with the maximum achievable accuracy and in a direct manner. For instance, Coordinates and Angles are direct measurements whereas areas, volumes etc. are indirectly calculated from other direct measurements. Therefore indices based on direct measurements are more accurate and should be preferable.
Indices should be independent from the method of measuring the back surface deformities. If this is not the case then indices can not be of universal use, and will also highly depend on the current technology.
Indices should be based on robust procedures and automatic measurements and should be evaluated by automatic processing techniques, eliminating as far as possible the human intervention. The reported levels of inter-/intra-observer variability and accuracy of the indices used so far reveals this problem. Only with automation the observer variability, the human induced errors, objectivity, and required experience will be eliminated.
Indices should be based on automatically detectable and uniquely identifiable anatomical landmarks. This is closely connected to point No. 4 above. Both the landmarks used and the measured points on the back surface should be unambiguously positioned, properly signalized and automatically detected and measured on the image.
Indices should require simple measuring protocols. Complicated or demanding protocols are sources of errors. This includes also (and especially) patient position and orientation relative to the sensor, lighting conditions, etc. Indices should be independent from and robust with respect to these parameters as much as possible.
Indices should be normalized in order to be comparable among patients. This means that the indices should not depend on the trunk size, on the width of the waist or the length of the arms. In this respect, indices should be unitless, percentages etc.
Indices should provide a stable datum for progress monitoring over time. This means that indices should either be coordinate-system-free of refer to a coordinate system which is stable over time.
Indices should be able to distinguish between different types of surface deformities, ie. Coronal/Transverse/Sagittal, Left/Right semi-trunk, Thoracic/Thoraco-Lumbar/Lumbar, Single/Double curves.
Indices should provide a clear and safe difference in magnitude between normality and pathology, so that pathology can be safely distinguished and diagnosed. This actually means increased sensitivity and specificity. It also means that the indices should have small typical error relative to the smallest change (progression) we would like to detect.
- Vasiliadis E, Grivas TB: Quality of life after conservative treatment of adolescent idiopathic scoliosis. Stud Health Technol Inform. 2008, 135: 409-13.PubMedGoogle Scholar
- Pazos , Cheriet VF, Danserau J, Ronsky RF, Zernicke H, Labelle : Reliability of trunk shape measurements based on 3-D surface Reconstructions. Eur Spine J. 2007, 16: 1882-1891. 10.1007/s00586-007-0457-0.View ArticlePubMedPubMed CentralGoogle Scholar
- Nault , Marie-Lyne , Allard P, Hinse S, Le Blanc R, Caron O: Relations Between Standing Stability and Body Posture Parameters in Adolescent Idiopathic Scoliosis. SPINE. 2002, 27 (17): 1911-1917. 10.1097/00007632-200209010-00018.View ArticlePubMedGoogle Scholar
- Bunnell WP: Selective screening for scoliosis. Clin Orthop Relat Res. 2005, 40-45. 10.1097/01.blo.0000163242.92733.66.Google Scholar
- Adams W: Lectures on the pathology and treatment of lateral and other forms of curvature of the spine. 1882, London: ChurchillGoogle Scholar
- Grivas TB, Vasiliadis ES, Koufopoulos G, Segos D, Triantafyllopoulos G, Mouzakis V: Study of trunk asymmetry in normal children and adolescents. Scoliosis. 2006, 1: 19-10.1186/1748-7161-1-19.View ArticlePubMedPubMed CentralGoogle Scholar
- Grossman TW, Mazur JM, Cummings RJ: An evaluation of the Adams forward bend test and the scoliometer in a scoliosis school screening setting. J Pediatr Orthop. 1995, 15: 535-View ArticlePubMedGoogle Scholar
- Cote P, Kreitz BG, Cassidy JD: A study of the diagnostic accuracy and reliability of the Scoliometer and Adam's forward bend test. Spine. 1998, 23: 796-10.1097/00007632-199804010-00011.View ArticlePubMedGoogle Scholar
- Viviani GR, Budgell L, Dok C, Tugwell P: Assessment of accuracy of the scoliosis school screening examination. Am J Public Health. 1984, 74: 497-10.2105/AJPH.74.5.497.View ArticlePubMedPubMed CentralGoogle Scholar
- Scherl S, Philips W, Torchia M: Clinical, features evaluation and diagnosis of adolescent idiopathic scoliosis. UpToDate. 2008, [http://www.uptodate.com]Google Scholar
- Grivas TB, Wade MH, Negrini S, O'Brien JP, Maruyama T, Hawes MC, Rigo M, Weiss HR, Kotwicki T, Vasiliadis ES, Sulam LN, Neuhous T: SOSORT consensus paper: school screening for scoliosis. Where are we today? Scoliosis. 2007, 2: 17-PubMedGoogle Scholar
- Grivas TB, Vasiliadis ES, Mihas C, Savvidou O: The effect of growth on the correlation between the spinal and rib cage deformity: implications on idiopathic scoliosis pathogenesis. Scoliosis. 2007, 2: 11-10.1186/1748-7161-2-11.View ArticlePubMedPubMed CentralGoogle Scholar
- Amendt LE, Ause-Ellias KL, Eybers JL: Validity and reliability testing of the Scoliometer. Phys Ther. 1990, 70: 108-PubMedGoogle Scholar
- Goldberg CJ, Kaliszer M, Moore DP, Fogarty EE, Dowling FE: Surface, Topography Cobb Angles and cosmetic Change in Scoliosis. Spine. 2001, 26 (4): E55-E63. 10.1097/00007632-200102150-00005.View ArticlePubMedGoogle Scholar
- Newton PO, Wenger DR: Idiopathic scoliosis. Lovell and Winter's Pediatric Orthopaedics. Edited by: Morrissy, RT, Weinstein, SL. 2006, Lippincott Williams & Wilkins, Philadelphia, 693-6Google Scholar
- Cobb JR: Outline for the study of scoliosis. American Academy of Orthopaedic Surgeons, Instructional Course Lectures. 1948, St. Louis: C.V. Mosby, 261-275.Google Scholar
- Cassar-Pullicino VN, Eisenstein SM: Imaging in scoliosis: what, why and how?. Clin Radiol. 2002, 57: 543-10.1053/crad.2001.0909.View ArticlePubMedGoogle Scholar
- Dickson RA, Weinstein SL: Bracing (and screening)-yes or no?. J Bone Joint Surg Br. 1999, 81: 193-10.1302/0301-620X.81B2.9630.View ArticlePubMedGoogle Scholar
- Morrissy RG, Goldsmith G, Hall E, Kehl D, Cowie H: Measurement of the Cobb angle on radiographs of patients who have scoliosis. J Bone Joint Surg Am. 1990, 72: 320-7.PubMedGoogle Scholar
- Dang NR, Moreau MJ, Hill DL: Intra-observer reproducibility and interobserver reliability of the radiographic parameters in the Spinal Deformity Study Group's AIS Radiographic Measurement Manual. Spine. 2005, 30: 1064-10.1097/01.brs.0000160840.51621.6b.View ArticlePubMedGoogle Scholar
- Srinivasalu S, Modi H, SMehta S, Suh S-W, Chen T, Murun T: Cobb Angle Measurement of Scoliosis Using Computer Measurement of Digitally Acquired Radiographs-Intraobserver and Interobserver Variability. Asian Spine Journal. 2008, 2 (2): 90-93. 10.4184/asj.2008.2.2.90.View ArticlePubMedPubMed CentralGoogle Scholar
- Petit Y, Aubin CE, Labelle H: Three-dimensional imaging for the surgical treatment of idiopathic scoliosis in adolescents. Can J Surg. 2002, 45: 453-458.PubMedPubMed CentralGoogle Scholar
- Sanders JO, Polly DW, Cats-Baril W, Jones JA, Lenke LG, O'Brien MF, Richards BS, Sucato DJ: Analysis of patient and parent assessment of deformity in idiopathic scoliosis using the Walter-Reed Visual Assessment Scale. Spine. 2003, 28: 2158-2163. 10.1097/01.BRS.0000084629.97042.0B.View ArticlePubMedGoogle Scholar
- Bago J, Climent JM, Pineda S, Gilperez C: Further evaluation of the Walter Reed Visual Assessment Scale: correlation with curve pattern and radiological deformity. Scoliosis. 2007, 2: 12-10.1186/1748-7161-2-12.View ArticlePubMedPubMed CentralGoogle Scholar
- Pineda S, Bago J, Climent JM, Gilperez C: Validity of the Walter Reed Visual Assessment Scale to measure subjective perception of spine deformity in patients with idiopathic scoliosis. Scoliosis. 2006, 1: 18-10.1186/1748-7161-1-18.View ArticlePubMedPubMed CentralGoogle Scholar
- Kotwicki T, Kinel E, Stryla W, Szulc A: Discrepancy in clinical versus radiological parameters describing deformity due to brace treatment for moderate idiopathic scoliosis. Scoliosis. 2007, 2: 18-10.1186/1748-7161-2-18.View ArticlePubMedPubMed CentralGoogle Scholar
- Zaina F, Negrini S, Atanasio S: TRACE (Trunk Aesthetic Clinical Evaluation), a routine clinical tool to evaluate aesthetics in scoliosis patients: development from the Aesthetic Index (AI) and repeatability. Scoliosis. 2009, 4: 3-10.1186/1748-7161-4-3.View ArticlePubMedPubMed CentralGoogle Scholar
- Kotwicki T, Negrini S, Grivas Th, Rigo M, Maruyama T, Durmala J, Zaina F: Methodology of evaluation of morphology of the spine and the trunk in idiopathic scoliosis and other spinal deformities -6th SOSORT consensus paper. Scoliosis. 2009, 4: 26-10.1186/1748-7161-4-26.View ArticlePubMedPubMed CentralGoogle Scholar
- Takasaki H: Moiré Topography. Applied Optics. 1970, 9: 1457-1492. 10.1364/AO.9.001467.View ArticleGoogle Scholar
- Turner-Smith A, Harris JD: ISIS - An automated shape measurement and analysis System. Proc of 3rd Int Symposium on Surface Topography and Spinal Deformity, Sept. 27-28. Edited by: Harris JD, Turner-Smith AR. 1984, Oxford, Gustav Fischer, Verlag, 31-28. ISBN 3-437-10962-6Google Scholar
- Turner-Smith A: A television/computer three-dimensional surface shape measurement system. J Biomech. 1988, 21 (6): 515-529. 10.1016/0021-9290(88)90244-8.View ArticlePubMedGoogle Scholar
- Weisz I, Jefferson RJ, Turner-Smith AR, Houghton GR, Harris JD: ISIS scanning: a useful assessment technique in the management of scoliosis. Spine. 1988, 13: 405-8. 10.1097/00007632-198804000-00006.View ArticlePubMedGoogle Scholar
- Theologis T, Fairbank JCT, Turner-Smith A, Pantazopoulos Th: Early Detection of Progression in Adolescent Idiopathic Scoliosis by Measurement of Changes in Back Shape With the Integrated Shape Imaging System Scanner. SPINE. 1997, 22 (11): 1223-1227. 10.1097/00007632-199706010-00010.View ArticlePubMedGoogle Scholar
- Berryman F, Pynsent P, Fairbank J, Disney Sm: A new system for measuring 3D back shape in scoliosis. Eur Spine J. 2008, 17: 663-672. 10.1007/s00586-007-0581-x.View ArticlePubMedPubMed CentralGoogle Scholar
- Groves D, Curran P: An accurate, fast and cost effective method for the measurement of body shape and the assessment of spinal deformity. Proceedings of the 6th international symposium on surface topography and spinal deformity. Edited by: Alberti A, Drerup B, Hierholzer E. 1992, Gustav Fischer, StuttgartGoogle Scholar
- Thometz JG, Lamdan R, Liu XC, Lyon R: Relationship Between Quantec Measurement and Cobb Angle in Patients with Idiopathic Scoliosis. Journal of Pediatric Orthopaedics. 2000, 20 (4): 512-516. 10.1097/00004694-200007000-00017.PubMedGoogle Scholar
- Oxborrow N: Assessing the child with scoliosis: the role of surface topography. Arch Dis Child. 2000, 83: 453-455. 10.1136/adc.83.5.453.View ArticlePubMedPubMed CentralGoogle Scholar
- Gomes AS, Serra LA, Lage AS, Gomes A: Automated 360° profilometry of human trunk for spinal deformity analysis. Three-dimensional analysis of spinal deformities. Edited by: D'Amico M. 1995, Amsterdam, NL: IOS Press, 423-9.Google Scholar
- Sciandra J, De JC, Mauroy G, Rolet R, Kohler JP: Accurate and fast non-contact 3D acquisition of the whole trunk. Three-dimensional analysis of spinal deformities. Amsterdam NL, Technology and Informatics 15. Edited by: D'Amico M. 1995, Oxford IOS Press, 81-85.Google Scholar
- Poncet P, Delorme S, Ronsky JL, Dansereau J, Harder J, Clynch G: Reconstruction of laser-scanned 3D torso topography and stereo-radiographical spine and rib-cage geometry in scoliosis. Comp Meth Biomech Biomed Eng. 2000, 4: 59-75. 10.1080/10255840008907998.View ArticleGoogle Scholar
- Schmitz A, Gabel H, Weiss HR, Schmitt O: Anthropometric 3D-body scanning in idiopathic scoliosis. Z Orthop Ihre Grenzgeb. 2002, 140 (6): 632-6. 10.1055/s-2002-36045.View ArticlePubMedGoogle Scholar
- Suzuki S, Yamamuro T, Shikita J, Shimozo K, Iida H: Ultrasound measurement of vertebral rotation in idiopathic scoliosis. Journal of Bone and Joint Surgery. 1989, 71 (B): 252-255.PubMedGoogle Scholar
- Frobin W, Hierholzer E: Rasterstereography: a photogrammetric method for measurement of body surfaces. Photogrammetric Engineering and Remote Sensing. 1981, 47 (12): 1717-1724.Google Scholar
- Frobin W, Hierholzer E: Automatic Measurement of body surfaces using rasterstereograph. Photogrammetric Engineering and Remote Sensing. 1983, 49 (3): 377-384.Google Scholar
- Hill DL, Raso VJ, Durdle NG, Peterson AE: A video-based technique for trunk measurement. Surface Topography and Spinal Deformity. Edited by: Alberti A, Drerup B, Hierholzer E. 1992, Stuttgart: Gustav Fischer Verlag, 52-56.Google Scholar
- Sechidis L, Tsioukas V, Patias P: An automatic process for the extraction of the 3D model of a human back for scoliosis treatment. International Archives of Photogrammetry and Remote Sensing. 2000, 33 (Supplement B5): 113-118.Google Scholar
- Patias P, Stylianidis E, Pateraki M, Chrysanthou G, Contozis C, Zavitsanakis Th: 3D digital photogrammetric reconstructions for scoliosis screening. Proceedings of the ISPRS Com. V Symposium, Dresden, Germany, The International Archives of the Photogrammetry, Remote Sensing and Spatial Information Sciences. 2006, XXXVI (Part 5): ISSN 1682-1750Google Scholar
- Willner S: Moiré Topography: A method for school screening of scoliosis. Arch Orthop Trauma Surg. 1979, 95: 181-185. 10.1007/BF00460117.View ArticlePubMedGoogle Scholar
- Willner S: Moiré Topography for the diagnosis and documentation of scoliosis. Acta Orthop Scand. 1979, 50: 295-302. 10.3109/17453677908989770.View ArticlePubMedGoogle Scholar
- Sahlstrand T: The clinical value of Moiré Topography in the management of scoliosis. Spine. 1986, 11 (5): 409-417. 10.1097/00007632-198606000-00002.View ArticlePubMedGoogle Scholar
- Denton T, Randall F, Deinlein D: The use of instant Moiré Photographs to reduce exposure from scoliosis radiographs. Spine. 1992, 17 (5): 509-512. 10.1097/00007632-199205000-00007.View ArticlePubMedGoogle Scholar
- Grivas T, Karras G, Katrabasas J, Papavasiliou N: Study of posterior trunk surface changes by age and sex using moiré topography. Research into spinal deformities 1. Edited by: Sevastik J, Diab K. 1997, IOS Press, Amsterdam, 331-334.Google Scholar
- Bendels G, Klein R, Samini M, Schmitz A: Statistical shape analysis for computer aided spine deformity detection. Journal of WSCG. 2005, 13-Google Scholar
- Duong L, Mac-Thiong J-M, Labelle H: Real time non invasive assessment of external trunk geometry during surgical correction of adolescent idiopathic scoliosis. Scoliosis. 2009, 4: 5-10.1186/1748-7161-4-5.View ArticlePubMedPubMed CentralGoogle Scholar
- Thometz J, Liu X, Lyon R, Harris G: Variability in Three-Dimensional Measurements of Back Contour with Raster Stereography in Normal Subjects. Journal of Pediatric Orthopaedics. 2000, 20 (1): 54-10.1097/00004694-200001000-00012.PubMedGoogle Scholar
- Suzuki N, Inami K, Ono T, Kohno K, Asher M: Analysis of posterior trunk symmetry index (POTSI) in Scoliosis: Part 1. Research into Spinal Deformities 2. Edited by: Stokes IAF. 1999, Amsterdam: IOS Press, 81-4.Google Scholar
- Asher M, Min Lai S, Burton D, Manna B: The Influence of Spine and Trunk Deformity on Preoperative Idiopathic Scoliosis Patients' Health related Quality of Life Questionnaire Responses. SPINE. 2004, 29 (8): 861-868. 10.1097/00007632-200404150-00008.View ArticlePubMedGoogle Scholar
- Asher M, Min Lai S, Burton D, Manna B: Maintenance of Trunk Deformity Correction Following Posterior Instrumentation and Arthrodesis for Idiopathic Scoliosis. SPINE. 2004, 29 (16): 1782-1788. 10.1097/01.BRS.0000134568.45154.34.View ArticlePubMedGoogle Scholar
- Minguez MF, Buendía M, Cibrián RM, Salvador R, Laguía M, Martín A, Gomar F: Quantifier variables of the back surface deformity obtained with a noninvasive structured light method: evaluation of their usefulness in idiopathic scoliosis diagnosis. Eur Spine J. 2007, 16 (1): 73-82. 10.1007/s00586-006-0079-y.View ArticlePubMedGoogle Scholar
- Jaremko JL, Poncet P, Ronsky J, Harder J, Dansereau J, Labelle H, Zernicke RF: Indices of torso asymmetry related to spinal deformity in scoliosis. Clinical Biomechanics. 2002, 17: 559-568. 10.1016/S0268-0033(02)00099-2.View ArticlePubMedGoogle Scholar
- Inami K, Suzuki N, Ono Y, Yamashita Y, Morisue H: Analysis of posterior trunk symmetry (POTSI) in scoliosis. Part 2. Research into Spinal Deformities 2. Edited by: Stokes IAF. 1999, Amsterdam: IOS Press, 85-8.Google Scholar
- Bunnell WP: An objective criterion for scoliosis screening. Journal of Bone and Joint Surgery. 1984, 66A (9): 1381-1387.Google Scholar
- Burwell R, James N, Johnson F, Webb J, Wilson Y: Trunk asymmetry scores: A method and some findings for normal children. Proc. of the 2nd Int. Symposium on Surface Topography and Spinal Deformity. Edited by: Drerup B, Frobin W, Hierholzer E. 1982, Muenster, Gustav Fischer Verlag, 29-32. ISBN 0-89574-178-4Google Scholar
- Pearson JD: Automated Visual Measurement of body shape in scoliosis. PhD Dissertation. 1996, School of Electrical and Electronic Engineering, Liverpool John Moores UniversityGoogle Scholar
- Stagnara P: Ambulante orthopaedische behandlung der scheuer-mannschen krankheit im floriden stadium. Schweiz Med Wschr. 1965, 95: 674-Google Scholar
- Sinoto A, Ohtsuka Y, Inoue S: Three dimensional analysis of the effect of brace treatment on idiopathic scoliosis. Proc. of 4th Int. Symposium on Surface Topography and Spinal Deformity. Edited by: Stokes I, Pekelsky J, Moreland M. 1986, Mont S. Marie, Quebec, Gustav Fischer Verlag, 113-130. ISBN 0-89574-261-6Google Scholar
- Gallup B, Cotton C, Ssewchenko N, Filoso S, Hidson D: Evaluation of 3D human shape monitoring techniques. Proc. of the 5th Int. Symposium on Surface Topography and Body Deformity, Sept. 29 - Oct. 1, 1988, Wien, Gustav Fischer Verlag, ISBN 1-56081-303-2. Edited by: Neugebauer H, Windischbauer G. 1990, Wien, Gustav Fischer Verlag, 119-126.Google Scholar
- Suzuki N, Ono T, Tezuka M, Kamilshi S: Moire topography and back shape analysis-clinical application. International Symposium on 3-D Scoliotic Deformities. Edited by: Dansereau J. 1992, Stuttgart: Gustav Fisher Verlag, 124-8.Google Scholar
- Kotwicki T, Kinel E, Chowanska J, Bodnar-Nanus A: POTSI, Hump Sum and Sum of Rotation - new surface topography parameters for evaluation of scoliosis deformity of the trunk. 2008, Fizjoterapia Polska, Medsportpress, 3 (4): 231-240. Vol. 8Google Scholar
- Goldberg JC, Kalszer M, Moore DP, Fogarty EE, Dowling FE: Surface Topography, Cobb angle and cosmetic changes in scoliosis. Spine. 2001, 26: 55-63. 10.1097/00007632-200102150-00005.View ArticleGoogle Scholar
- Weiss HR: Measurement of vertebral rotation: Perdriolle versus Raimondi. Eur Spine J. 1995, 4 (1): 34-38. 10.1007/BF00298416.View ArticlePubMedGoogle Scholar
- Weiss HR, Verres C, Steffan K, Heckel I: Outcome Measurement of Scoliosis Rehabilitation by Use of Surface Topography. Research into Spinal Deformities II. Edited by: Stokes IAF. 1999, IOS Press, Amsterdam, 246-249.Google Scholar
- Weiss HR, Lohschmidt K, El Obeidi N: Trunk Deformity in Relation to Breathing. A Comparative Analysis with the Formetric System. Research into Spinal Deformities I. Edited by: Sevastik JA, Diab KM. 1997, IOS Press, Amsterdam, 323-326.Google Scholar
- Weiss HR, Lohschmidt K, El Obeidi N: The Automated Surface Measurement of the Trunk. Technical Error. Edited by: Sevastik JA, Diab KM. 1997, IOS Press, Amsterdam, 305-308.Google Scholar
- Weiss HR, Dieckmann J, Gerner H J: Outcome of In-patient Rehabilitation in Patients with M. Scheuermann Evaluated by Surface Topography. Research into Spinal Deformities III. Edited by: Tanguy A, Peuchot B. 2002, IOS Press, Amsterdam, 246-249.Google Scholar
- Weiss HR, Dieckmann Jorg, Gerner Jurgen Hans: The practical use of surface topography: following up patients with Scheuermann's disease. Pediatr Rehabil. 2003, 6 (1): 39-45.View ArticlePubMedGoogle Scholar
- Kotwicki T: Evaluation of scoliosis today: Examination, X-rays and Beyond in Disability & Rehabilitation. Disability and Rehabilitation. 2008, 30 (10): 742-751. 10.1080/09638280801889519.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.